The biomechanical study of the influence to the forefoot plantar pressure of the first tarsometatarsal joint fracture-dislocation fixed by three different implants.
Objectives: To study the influence of forefoot plantar pressure of the first tarsometatarsal joint fracture-
dislocation by three different implants to provide experimental reference in selecting implants.
Methods: Eight fresh foot specimens were made into the models of the first tarsometatarsal joint fracture dislocation, which were fixed with 3. 5 mm cortical screw, 1/4 tubular plate and compressive staple in turn. After the loading of 600N, the changes of the plantar pressure in forefoot were measured by the method of the F-scan plantar pressure system.
Results: After first tarsometatarsal joint fracture-dislocation, the peak pressure under the first metatarsal head would decrease, while the pressure under the second metatarsal head would increase(P<0. 05). When the first tarsometatarsal joint was fixed with screw or plate respectively; the peak pressure under the two metatarsal heads would tend to be normal. However, the staple fixation would show the statistical significant difference, although the peak pressure under the first and second metatarsal heads were recovered in some extent(P0. 05), however, significant difference of the peak pressure was found in the first metatarsal head group and the second metatarsal group (P<0. 05).
When the first TMT joint was injured (condition 2), the peak pressure in the first MH would decrease while the peak pressure in the second MH would increase compared with condition 1 (P<0. 05).
When the first TMT joint was fixed by screw (condition 3) and plate (condition 4) separately, the peak pressure in the first MH would increase while the peak pressure in the second MH would decrease compared with condition 2 (P<0. 05).
However, there were no significant differences compared the condition 3 with the condition 4 of the peak pressure in the first MH and the second MH(P<0. 05).
When the first TMT joint was fixed by staple (condition 5), although the peak pressure in the first MH would increase, it was still <the peak pressure in condition 1 (P<0. 05). On the contrary, although the peak pressure in the second MH would decrease, it was still more than the peak pressure in condition 1 (P<0. 05). While compared the condition 5 with the condition 3 and condition 4, peak pressure in the first MH in condition 5 is <condition 3 and condition 4, however, the peak pressure in the second MH in condition 5 is more than condition 3 and condition 4(P<0. 05).
The model establishment of the first TMT joint fracture-dislocation: The direction of the first TMT joint dislocation can be divided into dorsal dislocation and plantar dislocation in the cross section. In clinical, dorsal dislocation is more common than plantar dislocation. 5 Furthermore, midfoot injury often associated with ligaments injury and fracture in other parts of the foot, but the present experimental conditions cannot completely simulate specific parts of the fracture dislocation and ligament injury according to the injury mechanism. Therefore, in this study, the model of the first TMT joint fracture-dislocation could only be established by cutting the ligaments in the first TMT joint and doing the osteotomy in the base of the first metatarsal. This model could be classified as Type B1 of Myerson classification. In the normal gait cycle, the load of the TMT joint increased gradually from initial contact to terminal stance and it reached the maximum in the pre-swing.
At this time, the ankle was 15deg plantar flexion. However, if the gait was larger, the plantar flexion range would increase. The TMT joint was easy to be injured when the load was axially loaded to the ankle in an extreme plantar flexion state. Therefore, in this study the specimen was axially loaded in 30deg plantar flexion of the ankle with the maximum load of 600N (body weight of a normal adult). 6 Actually, in gait cycle, the TMT joint bears more than 600N, however, when the load exceeded 600N, the fixation of ankle in 30degplantar flexion would easily failed. The changes of forefoot plantar pressure of the first TMT joint fracture-dislocation: Plantar pressure measurement is a clinical detection technology to measure the static or dynamic plantar pressure to reveal the characteristics of plantar pressure distribution, which could provide references to analyzed the etiology, progression and judge the prognosis.
Schepers7 studied the gait and plantar pressure of 26 patients of TMT joints injury and the results showed that compared with the contralateral foot, the contact area and contact time of the affected forefoot to the ground was reduced, while the contact area and the peak pressure of midfoot increased. If this state could not be corrected in a long-term, it may become one of the causes of pain in midfoot. In this study, we first measured the forefoot plantar pressure of 600N before and after the first TMT joint injury. The results showed the injury of the integrity of the first TMT joint would cause the decrease of the peak pressure in the first MH and increase of the peak pressure in the second MH. We hypothesize that there might be a regulating mechanism of "load transfer" in plantar pressure of the forefoot. In normal condition, the plantar pressure in the first MH can be regulated by the fat pad and the plantar fascia. 8
However, the first TMT joint instability would affect the activity of the first metatarsal and the self-regulating mechanism might be impaired. The contact area of first MH to the ground would be increased and the buffer capacity of first MH to the plantar pressure would be reduced. The reduced part might need to be compensated by the adjacent hallux and the second MH. Although this kind of "load transfer" could relieve the plantar pressure in the injury of TMT joint to some extent, the additional increased plantar pressure in other part of the forefoot might cause the complications such as the forefoot metatarsalgia, painful callosities, osteonecrosis of the MH, plantar fasciitis and pressure ulcer if it existed in a long time. 9, 10
The influence to the forefoot plantar pressure of the first TMT joint fracture-dislocation fixed by three different implants: Currently, it is emphasized that in the process of the fracture reduction or the correction of the complicated foot deformity, the restoration of the plantar pressure balance is more important than the foot shape recovery. 11 Schepers12 studied the plantar pressure in 21 patients of intra-articular calcaneal fracture and concluded that there was no direct relationship between the plantar pressure distribution and the clinical effect. Therefore, the authors believed that the clinical evaluation cannot completely replace the plantar pressure analysis. At present, the plantar pressure analysis is often used to determine the appropriate implants and evaluation of the surgical efficacy. 13
In this study, the results showed that the screw or plate fixation to the first TMT joint could make the peak pressure in the first MH and the second MH tend to be normal. There was no significant difference in the ability of the plate and screw to restore the peak pressure to be normal in the first MH and the second MH. When the first TMT joint was fixed by the staple, although the peak pressure would decrease in the second MH and increase in the first MH, significant differences could still be found compared with the normal state and the fixation by screw and plate. The results illustrated that when the stability of the first TMT joint was restored, the additional load born by hallux and the second metatarsal would be redistributed again to the first MH by the "load transfer" mechanism, the forefoot plantar pressure will therefore be balanced.
It suggests that when the first TMT joint is fracture dislocation, the accurate anatomical reduction and internal fixation by screw or plate helps to maintain the balance of the plantar pressure to avoid foot disease. As for the staple, we speculated that the staple fixation in complete dislocation of TMT joint (both dorsal and plantar ligaments injury) may cause fixation failure because of the less pullout resistance of the staple in this type of injury. Linked to this study, if the specimen of first TMT joint fracture dislocation was fixed by staple, the stability of the first TMT joint could only be recovered partly. These results suggested that attention should be paid to indications of the clinical application of staple.
This study was supported by a grant from the Natural Science Foundation of Ningbo City (No. 2014A610250) and Rising Star in Medicine Project in Zhejiang Province (No. 2015-70).
1. Arastu MH, Buckley RE. Tarsometatarsal joint complex and midtarsal injuries. Acta Chir Orthop Traumatol Cech. 2012;79(1):21-30.
2. Russell NA, Regazzola G, Aiyer A, Nomura T, Pelletier MH, Myerson M, et al. Evaluation of Nitinol Staples for the Lapidus Arthrodesis in a Reproducible Biomechanical Model. Front Surg. 2015, 14;2:65. doi: 10. 3389/fsurg. 2015. 00065. eCollection 2015.
3. Chevalier TL, Hodgins H, Chockalingam N. Plantar pressure measurements using an in-shoe system and a pressure platform: a comparison. Gait Posture. 2010, 31(3):397-399. doi: 10. 1016/j. gaitpost. 2009. 11. 016.
4. Alberta FG, Aronow MS, Barrero M, Diaz-Doran V, Sullivan RJ, Adams DJ. Ligamentous Lisfranc joint injuries: a biomechanical comparison of dorsal plate and transarticular screw fixation. Foot Ankle Int. 2005;26(6):462-473.
5. Kitsukawa K, Hirano T, Niki H, Tachizawa N, Nakajima Y, Hirata K. MR Imaging Evaluation of the Lisfranc Ligament in Cadaveric Feet and Patients With Acute to Chronic Lisfranc Injury. Foot Ankle Int. 2015;36(12):1483-1492. doi: 10. 1177/1071100715596746.
6. Panchbhavi VK, Andersen CR, Vallurupalli S, Yang J. A minimally disruptive model and three-dimensional evaluation of Lisfranc joint diastasis. J Bone Joint Surg Am. 2008;90(12):2707-2713. doi: 10. 2106/JBJS. G. 01420.
7. Schepers T, Kieboom B, van Diggele P, Patka P, Van Lieshout EM. Pedobarographic analysis and quality of life after Lisfranc fracture dislocation. Foot Ankle Int. 2010;31(10):857-864. doi: 10. 3113/FAI. 2010. 0857.
8. Ko PH, Hsiao TY, Kang JH, Wang TG, Shau YW, Wang CL. Relationship between plantar pressure and soft tissue strain under metatarsal heads with different heel heights. Foot Ankle Int. 2009;30(11):1111-1116. doi: 10. 3113/FAI. 2009. 1111.
9. Willegger M, Holinka J, Ristl R, Wanivenhaus AH, Windhager R, Schuh R. Correction power and complications of first tarsometatarsal joint arthrodesis for hallux valgus deformity. Int Orthop. 2015;39(3):467-476. doi: 10. 1007/s00264-014-2601-x.
10. Peterson KS, McAlister JE, Hyer CF, Thompson J. Symptomatic Hardware Removal After First Tarsometatarsal Arthrodesis. J Foot Ankle Surg. 2016;55(1):55-59. doi: 10. 1053/j. jfas. 2015. 06. 001.
11. Eleftheriou K I, Rosenfeld P F, Calder J D. Lisfranc injuries: an update. Knee Surg Sports Traumatol Arthrosc. 2013;21(6):1434-1446. doi: 10. 1007/s00167-013-2491-2.
12. Schepers T, Van der Stoep A, Van der Avert H, Van Lieshout EM, Patka P. Plantar pressure analysis after percutaneous repair of displaced intra-articular calcaneal fractures. Foot Ankle Int. 2008;29(2):128-135. doi: 10. 3113/FAI. 2008. 0128.
13. Schuh R, Hofstaetter JG, Hofstaetter SG, Adams SB Jr, Kristen KH, Trnka HJ. Plantar pressure distribution after tibiotalar arthrodesis. Clin Biomech (Bristol Avon). 2011;26(6):620-625. doi: 10. 1016/j. clinbiomech. 2011. 02. 001.
|Printer friendly Cite/link Email Feedback|
|Publication:||Pakistan Journal of Medical Sciences|
|Date:||Feb 28, 2017|
|Previous Article:||Prevalence and causes of visual impairment among Saudi adults.|
|Next Article:||Cognitive dysfunction in patients with Systemic Lupus Erythematosus.|