Software problems leading to false-positive results.
In the study, 259 patients underwent myocardial perfusion PET-CT using a 16-slice scanner. The researchers measured misregistration of superimposed PET-CT fusion images in millimeters and correlated the inaccuracies with associated quantitative size and severity of PET defects.
There were false-positive results in 40 percent of the 259 patients. In 23 percent of those patients, the results showed severe false abnormalities suggesting heart disease, which researchers said could have resulted in unwarranted emergency heart procedures if they had not run additional tests to verify the accuracy of the results.
The problem was attributed to an erroneous basic concept in the software functions of the PET-CT scanners. The commercial software does not account for movement in the lungs and heart while the patient is breathing during the test, and as a result, the PET and CT images don't always match or coregister.
Researchers developed a solution by rewriting the software to account for breathing motions and to correctly align PET and CT images. However, the redesigned software is not available commercially at this time.
|Printer friendly Cite/link Email Feedback|
|Title Annotation:||in the news|
|Article Type:||Brief article|
|Date:||Oct 1, 2007|
|Previous Article:||Report questions prenatal ultrasound screening.|
|Next Article:||New blog focuses on seniors with cancer.|
|False-Positive HIV Results. (Feature Article).|