Recommendations for improving U.S. NSF-supported airborne microwave radiometry.
These considerations motivated a workshop on airborne radiometry for water vapor and liquid water retrievals held at the National Center for Atmospheric Research (NCAR) on 23-24 September 2014. The workshop provided a critical opportunity to revisit U.S. NSF capabilities, forge a community consensus on scientific requirements, and discuss infrastructure and institutional resources. These discussions culminated in a set of technical and institutional process recommendations.
The science goals were distinguished through two foci. The first is a critical need for path-integrated liquid water and water vapor measurements for clouds occupying well-mixed layers, possibly in conditions cold enough to support mixed-phase microphysics. Such cloud characterization, useful in itself, also geophysically constrains retrievals of cloud vertical structure from active sensors. Multifrequency measurements deliver robust retrievals within a range of humidity environments, spanning the low latitudes to the Arctic. The second focus is on three-dimensional humidity mapping, driven by mesoscale research into Earth's water cycle but also applicable to climate, where new understanding is linking moisture transport to the low cloud feedback. Specific science goals include understanding the humidity structure of hurricanes and atmospheric rivers, and the marine boundary layer and its overlying troposphere, understanding the relationship of humidity to convection including orographic redistributions, and improving cloud forecast prediction. Both foci highlight the free troposphere and below.
Microwave emission is illustrated from the vantage point of an aircraft radiometer flying at 0.5 and 1.2 km looking up in Fig. 1, and at 12 km looking down in Fig. 2. The opacity of the cloudy atmosphere increases with frequency as a quasiquadratic function of liquid water path, punctuated by water vapor and oxygen absorption bands (Fig. 1). Sampling about the water vapor bands provides integrated water vapor and liquid water measurements, with measurements between 20 and 30 GHz most useful within moist environments because the absorption band is too weak to saturate, and measurements around 183 GHz suited to dry conditions where the strong sensitivity of the measurements to water vapor can be exploited. Figure 2 illustrates weighting functions (the sensitivity of microwave transmission to the vertical distribution of the water vapor, from which atmospheric humidity profiles can be produced) around 183 GHz. These frequencies are used by satellites as well (e.g., the Advanced Microwave Sounder Unit), in part because higher-frequency channels offer higher resolution and/or can be operated with smaller antennas.
BEST PRACTICES INSTRUMENT MENTORSHIP. After careful discussion, participants concluded that the calibration and retrieval development is best done by the scientists themselves, rather than by the vendor or developer. This is because vendors differ in their calibration approaches, and a consensus on the best approach is still lacking, with information sharing limited by patent rights in some cases. Instead, calibration and subsequent model evaluation is best achieved by comparison to the many DOE radiometers at its Southern Great Plains site because of DOE's long history of evaluating multiple-vendor instruments. The retrieval development and evaluation by scientists is also not hampered by proprietary software. Overall, scientists are best motivated to develop and document retrievals meeting standards of accuracy, reproducibility, and independent verification. Such motivations, for example, encourage retrievals that incorporate the latest gaseous absorption coefficients, still an area of active research at the higher frequencies.
The needed retrieval products can also be divided into the integrated amounts (liquid water and water vapor) and profiles of humidity. In both cases, the challenge is to determine a process for making the best performing radiometer available to the research community. Once a requestable instrument is available for path-integrated measurements, responsibility for it and its retrievals could reside with a university scientist as well as within NCAR. Airborne radiometers capable of humidity profiling or scanning require more support. The retrieval development of profiles requires more sophistication than of path-integrated values. Retrieval methods can in theory be developed from early, dedicated test flight datasets and provided as part of a subsequent campaign instrument request. The initial retrieval development, not linked to any particular campaign, would most likely require collaboration between university or government laboratory scientists, instrument developers, and NCAR scientists (or else need to motivate new activity within NCAR).
RECOMMENDATIONS. Technological improvements are producing passive radiometer designs that enhance the overall synergism of airborne measurements, providing new insights into Earth's cloudy atmosphere. No radiometers capable of sensing any of water's phases are available within the NSF deployment pool. We recommend against the leasing arrangements that satisfy short-term campaign requirements, as they do not encourage the development of an optimal radiometer and do not provide good instrument mentorship, nor support practices best suited for scientific advancement. Instead, we recommend additional investment in microwave and millimeter-wavelength radiometers in the United States toward providing critical cloud and atmospheric characterization and to support a future three-dimensional thermodynamic mapping capability.
Radiometers emphasizing a compact design should fit into a standard cloud probe canister. Irrespective of size, radiometers should possess
* multifrequency humidity and liquid water sensing (20-30, 90, 183 GHz) capability;
* modularity of frequency components;
* the ability to be deployed upon multiple aircraft;
* views above and below the aircraft;
* complete documentation of internal measurements and calibration procedures;
* vendor-independent calibration and retrieval development plan; and,
* ground-based deployability to aid radiometer assessment at the Southern Great Plains DOE Atmospheric Radiation Measurement site.
Processes by which NSF might promote the development and/or acquisition of a new sensor include a joint development effort between NCAR and another entity (university, other government laboratory, or private-sector company) that already possesses the development capability. If a prototype sensor of interest already exists, we encourage further investment to develop the appropriate technical readiness for scientific fieldwork. This could include an expansion of NCAR's Airborne Research Instrumentation Testing Opportunity (ARISTO) program to include vendor-independent calibration and retrieval development. Sensor maintenance, calibration, operation, and retrieval development could occur at either NCAR, another government laboratory, or at a university, and follow "best practices" for instrument mentorship.
ADDITIONAL RESOURCES. The full work shop report is available at www.eol.ucar.edu/content /workshop-airborne-radiometry-water-vapor-and -liquid-water-retrievals. Measurement integration based on mixed-layer cloud science is discussed in Wood et al. (2011), Wang et al. (2012), and Zuidema et al. (2012). Further background on available temperature profiling capabilities can be found in Haggerty et al. (2014), while German capabilities in free-tropospheric thermodynamic profiling are detailed in Mech et al. (2014). An example of NASA high-altitude thermodynamic profiling integrated with active microwave sensing and in situ measurements to understand hurricane evolution is given in Braun et al. (2013). A previous NSF workshop report emphasizing the value of humidity profiling can be found at www .eol.ucar.edu/system/files/LAOF_Workshop _Report_FINAL_06112013.pdf.
ACKNOWLEDGMENTS. We thank the NSF Lower Atmospheric Observing Facilities Division and NCAR Earth Observing Laboratory for supporting the workshop on airborne microwave radiometry. Three anonymous reviewers and Ian McAdams of the Naval Research Laboratory provided comments that clarified and improved the final document.
FOR FURTHER READING
Braun, S. A., and Coauthors, 2013: NASAs Genesis and Rapid Intensification Processes (GRIP) field experiment. Bull. Amer. Meteor. Soc., 94, 345-363, doi:10.1175/bams-d-11-00232.1.
Haggerty, J., K. Schick, M. Mahoney, and B. Lim, 2014: The NCAR microwave temperature profiler: Data applications from recent deployments. Proc., 13th Specialist Meeting on Microwave Radiometry and Remote Sensing of the Environment, Pasadena, CA.
Mech, M., E. Orlandi, S. Crewell, F, Ament, L. Hirsch, M. Hagen, G. Peters, and B. Stevens, 2014: HAMP the microwave package on the high altitude and long range research aircraft HALO. Atmos. Meas. Tech., 7, 4539-4553, doi:10.5194/amt-7-4539-2014.
Wang, Z., and Coauthors, 2012: Single aircraft integration of remote sensing and in situ sampling for the study of cloud microphysics and dynamics. Bull. Amer. Meteor. Soc., 93, 653-668, doi:10.1175/BAMS -D-11-00044.1.
Wood, R., C. S. Bretherton, D. Leon, A. D. Clarke, P. Zuidema, G. Allen, and H. Coe, 2011: An aircraft case study of the spatial transition from closed to open mesoscale cellular convection over the southeast pacific. Atmos. Chem. Phys., 11,2341-2370, doi:10.5194 /acp-11-2341-2011.
Zuidema, P., D. Leon, A. Pazmany, and M. Cadeddu, 2012: Aircraft millimeter-wave passive sensing of cloud liquid water and water vapor during VOCALSREx. Atmos. Chem. Phys., 12, 355-369, doi:10.5194 /acp-12-355-2012.
AFFILIATIONS: Zuidema--University of Miami, Miami, Florida; Haggerty, Jensen, and Vivekanandan--National Center for Atmospheric Research, Boulder, Colorado; Cadeddu--Argonne National Laboratory, Argonne, Illinois; Orlandi and Mech--University of Cologne, Cologne, Germany; Wang--University of Wyoming, Laramie, Wyoming
CORRESPONDING AUTHOR: Paquita Zuidema, Rosenstiel School of Marine and Atmospheric Science, University of Miami, Miami, FL, 33149
TABLE 1. NSF aircraft deployment pool: active and passive remote sensing systems. Instrument Frequency Platform (wavelength) Active Wyoming Cloud Radar (WCR) 94 GHz (3.2 mm) King Air, C-130 Wyoming Cloud Lidar (WCL) (355 nm) King Air, C-130 HIAPER (1) Cloud Radar 94 GHz (3.2 mm) Gulfstream-V (HCR) High Spectral Resolution (532 nm) Gulfstream-V Lidar (HSRL) Multi-function Airborne (266,355 nm) King Air, C-130 Raman Lidar (MARLi) (2) Passive Microwave Temperature 56-59 GHZ Gulfstream-V Profiler (MTP) HIAPER1 Airborne (300-2400 nm) Gulfstream-V Radiation Package Kipp and Zonen pyrgeo-, (4.5-40,0.2-3.6 pm) C-130, Gulfstream-V pyranometers Instrument Reference Active Wyoming Cloud Radar (WCR) http://flights.uwyo.edu/uwka/wcr/ Wyoming Cloud Lidar (WCL) http://flights.uwyo.edu/uwka/wcl/ HIAPER (1) Cloud Radar http://dx.doi.org/10.5065/D6BP00TP (HCR) High Spectral Resolution http://dx.doi.org/10.5065/D67W6976 Lidar (HSRL) Multi-function Airborne Raman Lidar (MARLi) (2) Passive Microwave Temperature www.eol.ucar.edu/instruments Profiler (MTP) /microwave-temperature-profiler HIAPER1 Airborne www.eol.ucar.edu/node/228 Radiation Package Kipp and Zonen pyrgeo-, www.eol.ucar.edu/node/232 pyranometers (1) High-performance Instrumented Airborne Platform for Environmental Research (2) The MARLi is not strictly a member of the NSF deployment pool, but was designed for two platforms that are, and can be incorporated into NSF campaigns through an individual proposal.
|Printer friendly Cite/link Email Feedback|
|Title Annotation:||United States. National Science Foundation|
|Author:||Zuidema, Paquita; Haggerty, Julie; Cadeddu, Maria; Jensen, Jorgen; Orlandi, Emiliano; Mech, Mario; V|
|Publication:||Bulletin of the American Meteorological Society|
|Date:||Dec 1, 2016|
|Previous Article:||When El Nino rages: how satellite data can help water-stressed islands.|
|Next Article:||"The stippling shows statistically significant grid points": how research results are routinely overstated and overinterpreted, and what to do about...|