Rapid Determination of Six Low Molecular Carbonyl Compounds in Tobacco Smoke by the APCI-MS/MS Coupled to Data Mining.
Low molecular carbonyl compounds, existing in the cigarette smoke and automobile exhaust, aroused extensive attention due to their adverse effects on human health [1,2]. Due to the relatively low boiling points and high vapor pressures, low molecular carbonyl compounds were easy to inhale by the human body and cause irritation of the eyes and throat, headache, sickness, and even cancers [3,4]. Carbonyl compounds, such as acetaldehyde, were listed as air toxics in the Clean Air Act Amendments of 1990. Moreover, the Texas Commission on Environmental Quality (TCEQ) developed inhalation toxicity factors for the evaluation of acrolein and crotonaldehyde concentrations in 2014. In conventional method, the analysis of carbonyl compounds was based on derivatization. Carbonyl compounds reacted to derivatization reagents firstly and then analysis by capillary electrophoresis, chromatography, or chromatography coupled with mass spectrometer [5-9]. However, these analytical methods usually involved derivatization and multistep sample preparation procedure, which were labor-intensive, time-consuming, and tedious.
Mass spectrometry is one of the extensively used techniques for the identification of molecules in complex mixture owing to its universality and accuracy [10-12]. Due to less fragment ion, several online and real-time analyses were established based on the combination of soft ionization source and mass spectrometer, which had been proven as a powerful analytical technique and applied in food safety, air pollution, and biology [13-16]. Atmospheric pressure chemical ionization (APCI) source is one of the important soft ionization sources [17,18]. The ionization of sample for APCI was carried out through ion-molecule reaction in gas phase, which was more suitable for the direct detection of gas sample. Recently, APCI-MS had been applied in direct monitoring of volatile organic compounds in ambient air [19-21].
In previous study the commercial APCI source of Xevo[TM] TQ MS (Waters) was modified to introduce sample directly into the ionization region, and acrolein and crotonaldehyde were investigated by the APCI-MS/MS. However, the isomers in low molecular carbonyl compounds such as acetone/propionaldehyde and butanone/butyraldehyde were difficult to analyze by the APCI-MS/MS due to the same molecular mass. To overcome the difficulty from the isomer, the MS/MS spectra under a series of CID energies were investigated to provide more information, which was in favor of differentiation through data mining. In this study, a method was developed for the rapid determination of low molecular carbonyl compounds by APCI-MS/MS. The quantitation procedure of six low molecular carbonyl compounds was performed with the characteristic ion of [[C[H.sub.3]O].sup.+] under CID energy of 5 and 15 eV. Finally, six low molecular carbonyl compounds in tobacco smoke were detected with APCI-MS/MS to test and prove the developed method.
2.1. Chemicals and Reagents. Acrolein, propionaldehyde, crotonaldehyde, and butyraldehyde (99.5%) were purchased from Quality Control Chemicals INC (USA); acetone and butanone (>99%) were purchased from Fisher (UK); deionized/distilled water was made by Thermo Scientific GenPure. Nitrogen gas was purchased from Yuanzheng Technology Company (China). Standards solution was prepared by diluting the 5000 [micro]g/mL stock solutions by water to a final concentration. 10 [micro]L solution was injected into the container to make gas sample with 0.01~5 [micro]g/L. Tobacco smoke (3R4F) was extracted from the tube behind the Cambridge filter about 30 cm on smoking machine provided by Zhengzhou Tobacco Research Institute (China).
2.2. Apparatus. To achieve the quantitative analysis of low molecular carbonyl compounds, the standards gas was made by the gasification of standards solution in a 2 L glass container as described in our previous study . In order to clean the container automatically, two solenoid valves were used to control the gas switching, and the flow rate of cleaning gas was 180 L/h.
In order to introduce gas sample directly into the ionization region, APCI source of Xevo[TM] TQ MS (Waters, USA) was modified and the schematic diagram was shown in Figure 1. The deactivated metal capillary (0.021" inner diameter, 0.029" outer diameter, Restek, USA) was sheathed by a metal tube with 1 mm inner diameter. High-pressure nitrogen with 0.6 MPa was passed between capillary and metal tube. As nitrogen passed the end of the capillary, the cross section of gas flow was increased abruptly, and a negative pressure was produced due to the Venturi effect. Thus, the sample was introduced into the mass spectrometer ionization region through capillary. The gas flow rate in capillary was about 110 mL/min. To decline the memory effects, the capillary was cleaned by the gas in the container before the sampling. Meanwhile, the sampling volume was about 220 ml (2 min, 110 ml/min), which was larger than that of dead volume in the capillary.
The operating conditions of mass spectrometry were as follows. Ionization mode was positive; discharge current was 4 [micro]A; APCI probe temperature was 100[degrees]C; desolvation gas flow was 150 L/h; cone gas flow was 10 L/h; collision gas flow rate was 0.25 mL/min.
3. Results and Discussion
3.1. Ionization of Six Low Molecular Carbonyl Compounds. The full scan mass spectra for the mixture of four aldehydes and mixture of two ketones with 0.5 [micro]g/L were displayed in Figure 2, respectively. Peaks at m/z 57, 59, 71, and 73 were observed in Figure 2(a), which were corresponding to the [[MH].sup.+] of acrolein, propionaldehyde, crotonaldehyde, and butyraldehyde, respectively. Peaks at m/z 59 and 73 were observed in Figure 2(b), which were corresponding to the [[MH].sup.+] of acetone and butanone, respectively. The generation of [[MH].sup.+] was ascribed to proton transfer reaction, where the proton affinities (PA) of low molecular carbonyl compounds were higher than that of water. In Figure 2(a), peak at m/z 55 was ascribed to [C.sub.4][H.sub.7.sup.+], which was produced by the loss of [H.sub.2]O from the protonated butyraldehyde.
Daughter scan mode was usually used to find characteristic fragment ions for quantitative analysis. The daughter scan mass spectra of low molecular carbonyl compounds under CID energy of 15 eV were shown in Figure 3. As for low molecular carbonyl compounds, the carbonyl group was activated by the ionization through proton transfer reaction, and [H.sub.2]O (m/z 57 [right arrow] 39, m/z 59 [right arrow] 41, m/z 71 [right arrow] 53, and m/z 73 [right arrow] 55) or CO (m/z 57 [right arrow] 29, m/z 71 [right arrow] 43, and m/z 73 [right arrow] 45) was easy to lose in the collision cell [22,23]. As for acetone/propionaldehyde, although the protonated aldehydes were easy to lose CO, the formation of m/z 31 was not likely produced through the loss of CO due to the instability of [C.sub.2][H.sub.7.sup.+] (m/z 31) in Figures 3(b) and 3(e). Therefore, the ion of m/z 31 was ascribed to [[C[H.sub.3]O].sup.+] which was formed by the C-C band break. In Figure 3, m/z 31 was observed in all the daughter scan mass spectra of six low molecular carbonyl compounds, which was ascribed to [[C[H.sub.3]O].sup.+]. Due to the relation of [[C[H.sub.3]O].sup.+] with carbonyl group, [[C[H.sub.3]O].sup.+] was used as characteristic ion for quantification.
3.2. Quantification of Isomer Carbonyl Compounds. Due to the same molecular mass, it was difficult to quantify the isomer carbonyl compounds in the mixture without the chromatographic separation. To overcome the quantitative difficulties, more structure information should be obtained to distinguish acetone/propionaldehyde and butanone/butyraldehyde. Due to the different functional groups or positions in the molecule, the analysis of carbonyl isomers could be performed by chromatography coupled with derivatization [24,25]. However, derivatization methods usually involved multistep sample preparation and long analytical period. MS/MS was usually used to obtain structure information under the selected CID energy, while the plentiful structure information reflected by the CID energy was not in full use. Therefore, the influence of CID energy on the MS/MS spectra was investigated under a series of CID energies, such as 5, 15, 25, and 35 eV. When the CID energy was at 5 eV, there were obvious differences of the fragment intensities (m/z 31) in Figure 4 for acetone, propionaldehyde, butanone, and butyraldehyde, which was in favor of differentiation through data mining. As the CID energy was increased to 25 eV, the actual intensities were decreased. Meanwhile, the fragment ion distribution and actual intensity of acetone and butanone were similar to those of propionaldehyde and butyraldehyde. Due to the marked differences in MS/MS spectra for isomers of aldehydes and ketones under 5 and 15 eV, detailed statistical evaluation of data is not needed for discrimination of propionaldehyde/acetone and butyraldehyde/butanone. Therefore, the CID energies of 5 and 15 eV were chosen for quantification of the six low molecule carbonyl compounds.
In Figure 4(a), peaks at m/z 31 of propionaldehyde were observed, while this of acetone in Figure 4(b) was markedly lower compared with that of propionaldehyde. When the CID energy was increased to 15 eV, the actual intensity of m/z 31 for acetone was increased about 1.6 times in Figures 3(e) and 4(b), while that of propionaldehyde was decreased about 86% in Figures 3(b) and 4(a). When the CID energy was increased from 5 to 15 eV, the actual intensity of m/z 31 for butanone in Figures 3(f) and 4(d) was increased about 5.8 times, while that of butyraldehyde was decreased about 25% in Figures 3(d) and 4(c).
The relationship between the ion intensity and the concentration was expressed as I = [sigma] x C. Here, I was the ion intensity, C was the ion concentration, and a was the response coefficient. The relation between ion intensity at m/z 31 and the isomer concentration under 5 and 15 eV could be expressed as follows:
[mathematical expression not reproducible]. (1)
Here, [[sigma].sub.1](n) and [[sigma].sub.2](n) were the response coefficient of m/z 31 under the CID energy of n for two isomers, respectively; [C.sub.1] and [C.sub.2] were the concentration of the two isomers in the mixture. I(n) was the measured ion intensity at m/z 31 for the mixture under the CID energy of n in the mass spectra.
3.3. Linear Calibration Curves. Table 1 showed the calibration curves function of six low molecular carbonyl compounds by daughter scan mode with fragment ion [[C[H.sub.3]O].sup.+] under the CID energy of 5 and 15 eV. With the use of the standard definition of S/N = 3, the detection limit of butyraldehyde was up to 0.003 [micro]g/L. Meanwhile, Figure S1 of supplementary information available online at https://doi.org/10.1155/2017/8260860 showed the calibration curves with the concentration of 0.01, 0.025, 0.05, 0.1, 0.5, 1.0, 2.5, and 5 [micro]g/L. It showed a good linear relationship between the ion intensity and concentration, and the response coefficients were extracted from the slope of the calibration curve.
Acetone and propionaldehyde mixture (1: 1) with 0.5 [micro]g/L in the container was prepared using the weighting method to test the reliability of quantification based on (1). The errors between the concentration calculated by (1) and the actual concentration calculated by weighting method ranged from -12% to 5% for acetone and from -8% to 3% for propionaldehyde. Moreover, butanone and butyraldehyde mixture (1: 1) with 0.5 [micro]g/L was also measured. The errors between the concentration calculated by (1) and the actual concentration calculated by weighting method ranged from -3% to 1% for butanone and from -10% to 2% for butyraldehyde, which demonstrated good accuracy achieved by (1).
3.4. Measurement of Six Low Molecular Carbonyl Compounds in Tobacco Smoke. Low molecular carbonyl compounds have been found as the combustion product of cigarette. In this study, 5 mL main stream smoke of cigarette extracted from the smoke machine was injected into the container, and 10 [micro]L water was injected into the container at the same time. After two minutes, the tobacco smoke was analyzed by the APCI-MS/MS. The full scan mass spectrum of tobacco smoke was shown in Figure 5; low molecular carbonyl compounds of acetaldehyde (m/z 45), acrolein (m/z 57), crotonaldehyde (m/z 71), acetone/propionaldehyde (m/z 59), and butanone/butyraldehyde (m/z 73) were observed clearly.
The daughter scan mass spectra of low molecular carbonyl compounds were shown in Figure S2 of supplementary information. According to the standards curve in Table 1, the value was calculated about 0.57 and 0.1 [micro]g/L for acrolein and crotonaldehyde in container. Based on the intensity change of [[C[H.sub.3]O].sup.+] under the CID energies of 5 and 15 eV, the values for acetone, propionaldehyde, butanone, and butyraldehyde in container were about 2.9, 0.49, 0.6, and 0.11 [micro]g/L through the calculation with (1). The contents of acrolein, acetone, propionaldehyde, crotonaldehyde, butanone, and butyraldehyde for a cigarette were about 63 [+ or -] 5.8,325 [+ or -] 82, 55 [+ or -] 9.7,11 [+ or -] 1.4, 67 [+ or -] 5.9, and 12 [+ or -] 1.8 [micro]g/cig (35 mL puff volume, 8 times), which agreed with the result of Wagner et al. and Sampson et al. [26,27]. The contents of acrolein and crotonaldehyde for a cigarette were different between this paper and  (57 and 9.5 [micro]g/cig); the error of the analysis was ascribed to two aspects. One is the instrument stability, which was about [+ or -]5%, and the other is the reproducibility of cigarette smoke produced by the combustion, which was about [+ or -]10%. The different result was ascribed to the statistical difference. The determination of six low molecular carbonyl compounds in tobacco smoke was less than four minutes, and the agreement of results further verified that the developed method was reliable for the determination of the low molecular carbonyl compounds.
Six low molecular carbonyl compounds of acrolein, acetone, propionaldehyde, crotonaldehyde, butanone, and butyraldehyde in the cigarette smoke were analyzed rapidly through the combination of APCI-MS/MS and data mining. As for the isomers of carbonyl compounds, the quantification was performed by the APCI-MS/MS under two CID energies of 5 and 15 eV. A quantitation procedure was developed for low molecular carbonyl compounds with characteristic ion of [CH3O]+. The capability of the developed method was demonstrated through quantification of six low molecular carbonyl compounds in tobacco smoke with analytical period less than four minutes.
Conflicts of Interest
The authors declare that there are no conflicts of interest regarding the publication of this paper.
The work was supported by National Natural Science Foundation of China (Grant no. 21675173).
 Y.-C. Lin, T.-Y. Wu, W.-C. Ou-Yang, and C.-B. Chen, "Reducing emissions of carbonyl compounds and regulated harmful matters from a heavy-duty diesel engine fueled with paraffinic/biodiesel blends at one low load steady-state condition," Atmospheric Environment, vol. 43, no. 16, pp. 2642-2647, 2009.
 S.-H. Jo and K.-H. Kim, "Development of a sampling method for carbonyl compounds released due to the use of electronic cigarettes and quantitation of their conversion from liquid to aerosol," Journal of Chromatography A, vol. 1429, pp. 369-373, 2016.
 M. Kitwattanavong, T. Prueksasit, D. Morknoy, T. Tunsaringkarn, and W. Siriwong, "Health risk assessment of petrol station workers in the inner city of Bangkok, Thailand, to the exposure to BTEX and carbonyl compounds by inhalation," Human and Ecological Risk Assessment, vol. 19, no. 6, pp. 1424-1439, 2013.
 H. Lu, J.-J. Tian, Q.-Y. Cai, S. Wen, Y. Liu, and N. Li, "Levels and health risk of carbonyl compounds in air of the library in Guangzhou, South China," Aerosol and Air Quality Research, vol. 16, no. 5, pp. 1234-1243, 2016.
 J. E. Szulejko and K.-H. Kim, "Derivatization techniques for determination of carbonyls in air," TrAC--Trends in Analytical Chemistry, vol. 64, pp. 29-41, 2015.
 W. Yuan, S. Li, and J. L. Edwards, "Extraction and quantitation of ketones and aldehydes from mammalian cells using fluorous tagging and capillary LC-MS," Analytical Chemistry, vol. 87, no. 15, pp. 7660-7666, 2015.
 J. M. Fernandez-Molina and M. Silva, "Micro solid-phase derivatization analysis of low-molecular mass aldehydes in treated water by micellar electrokinetic chromatography," Electrophoresis, vol. 35, no. 6, pp. 819-826, 2014.
 V. M. Da Silva, M. C. D. C. Veloso, E. T. Sousa et al., "Determination of 11 low-molecular-weight carbonyl compounds in marine algae by high-performance liquid chromatography," Journal of Chromatographic Science, vol. 44, no. 5, pp. 233-238, 2006.
 H.-J. C. Chen and W.-P. Lin, "Simultaneous quantification of 1,N2-propano-2'-deoxyguanosine adducts derived from acrolein and crotonaldehyde in human placenta and leukocytes by isotope dilution nanoflow LC nanospray ionization tandem mass spectrometry," Analytical Chemistry, vol. 81, no. 23, pp. 9812-9818, 2009.
 R. Boyd, "From physical chemistry to mass spectrometry to government lab manager in half a century," Mass Spectrometry Reviews, vol. 35, no. 2, pp. 272-310, 2016.
 H. Yu, J. J. Lu, W. Rao, and S. Liu, "Capitalizing resolving power of density gradient ultracentrifugation by freezing and precisely slicing centrifuged solution: enabling identification of complex proteins from mitochondria by matrix assisted laser desorption/ionization time-of-flight mass spectrometry," Journal of Analytical Methods in Chemistry, vol. 2016, Article ID 8183656, 7 pages, 2016.
 M. R. Bladergroen and Y. E. M. Van Der Burgt, "Solid-phase extraction strategies to surmount body fluid sample complexity in high-throughput mass spectrometry-based proteomics," Journal of Analytical Methods in Chemistry, vol. 2015, Article ID 250131, 2015.
 L. Hua, K. Hou, P. Chen et al., "Realization of in-source collision-induced dissociation in single-photon ionization time-of-flight mass spectrometry and its application for differentiation of isobaric compounds," Analytical Chemistry, vol. 87, no. 4, pp. 2427-2433, 2015.
 M. Wagner, D. Tonoli, E. Varesio, and G. Hopfgartner, "The use of mass spectrometry to analyze dried blood spots," Mass Spectrometry Reviews, vol. 35, no. 3, pp. 361-368, 2016.
 M.-Z. Huang, C.-H. Yuan, S.-C. Cheng, Y.-T. Cho, and J. Shiea, "Ambient ionization mass spectrometry," Annual Review of Analytical Chemistry, vol. 3, no. 1, pp. 43-65, 2010.
 T. R. Covey, B. A. Thomson, and B. B. Schneider, "Atmospheric pressure ion sources," Mass Spectrometry Reviews, vol. 28, no. 6, pp. 870-897, 2009.
 F. J. Andrade, J. T. Shelley, W. C. Wetze et al., "Atmospheric pressure chemical ionization source. 1. Ionization of compounds in the gas phase," Analytical Chemistry, vol. 80, no. 8, pp. 2646-2653, 2008.
 F. J. Andrade, J. T. Shelley, W. C. Wetzel et al., "Atmospheric pressure chemical ionization source. 2. Desorption-ionization for the direct analysis of solid compounds," Analytical Chemistry, vol. 80, no. 8, pp. 2654-2663, 2008.
 C.-Y. Jiang, S.-H. Sun, Q.-D. Zhang et al., "Fast determination of 3-ethenylpyridine as a marker of environmental tobacco smoke at trace level using direct atmospheric pressure chemical ionization tandem mass spectrometry," Atmospheric Environment, vol. 67, pp. 1-7, 2013.
 A. Okumura, Y. Takada, S. Watanabe et al., "Real-time air monitoring of mustard gas and lewisite 1 by detecting their in-line reaction products by atmospheric pressure chemical ionization ion trap tandem mass spectrometry with counterflow ion introduction," Analytical Chemistry, vol. 87, no. 2, pp. 1314-1322, 2015.
 W. Zhao, B. Lu, J. Liu et al., "Rapid determination of acrolein and crotonaldehyde in tobacco smoke with water-assisted APCIMS/MS," International Journal of Mass Spectrometry, vol. 409, pp. 16-20, 2016.
 P. Neta, Y. Simon-Manso, Y. Liang, and S. E. Stein, "Loss of H2 and CO from protonated aldehydes in electrospray ionization mass spectrometry," Rapid Communications in Mass Spectrometry, vol. 28, no. 17, pp. 1871-1882, 2014.
 D. Smith, T. W. E. Chippendale, and P. Spanel, "Reactions of the selected ion flow tube mass spectrometry reagent ions H3O+ and NO+ with a series of volatile aldehydes of biogenic significance," Rapid Communications in Mass Spectrometry, vol. 28, no. 17, pp. 1917-1928, 2014.
 D. Bourdin and V. Desauziers, "Development of SPME on-fiber derivatization for the sampling of formaldehyde and other carbonyl compounds in indoor air," Analytical and Bioanalytical Chemistry, vol. 406, no. 1, pp. 317-328, 2014.
 Y. Shigeri, T. Kamimura, M. Ando et al., "2-Hydrazinoquinoline: a reactive matrix for matrix-assisted laser desorption/ionization mass spectrometry to detect gaseous carbonyl compounds," European Journal of Mass Spectrometry, vol. 22, no. 2, pp. 83-90, 2016.
 K. Wagner, R. Higby, and K. Stutt, "Puff-by-puff analysis of selected mainstream smoke constituents in the kentucky reference 2R4F cigarette," Contributions to Tobacco Research, vol. 21, no. 5, pp. 273-279, 2014.
 M. M. Sampson, D. M. Chambers, D. Y. Pazo, F. Moliere, B. C. Blount, and C. H. Watson, "Simultaneous analysis of 22 volatile organic compounds in cigarette smoke using gas sampling bags for high-throughput solid-phase microextraction," Analytical Chemistry, vol. 86, no. 14, pp. 7088-7095, 2014.
Wuduo Zhao, (1,2) Qidong Zhang, (1) Binbin Lu, (1) Shihao Sun, (1) Shusheng Zhang, (2) and Jianxun Zhang (1)
(1) Zhengzhou Tobacco Research Institute, China National Tobacco Corporation, Zhengzhou 450001, China
(2) College of Chemistry and Molecular Engineering, Zhengzhou University, Zhengzhou 450001, China
Correspondence should be addressed to Shusheng Zhang; email@example.com and Jianxun Zhang; firstname.lastname@example.org
Received 31 October 2016; Revised 14 January 2017; Accepted 15 February 2017; Published 20 April 2017
Academic Editor: Serban C. Moldoveanu
Caption: Figure 1: Schematic diagram of modified APCI source.
Caption: Figure 2: (a) Full scan mass spectra of four aldehyde mixtures with 0.5 [micro]g/L for acrolein, propionaldehyde, crotonaldehyde, and butyraldehyde, respectively. (b) Full scan mass spectra of two ketone mixtures with 0.5 [micro]g/L for acetone and butanone, respectively.
Caption: Figure 3: Daughter scan mass spectra of six low molecular carbonyl compounds with 0.5 [micro]g/L under the CID energy of 15 eV (a) Acrolein, (b) propionaldehyde, (c) crotonaldehyde, (d) butyraldehyde, (e) acetone, and (f) butanone.
Caption: Figure 4: Daughter scan mass spectra of 0.5 [micro]g/L isomers of low molecular carbonyl compounds under the CID of 5 eV. (a) Propionaldehyde, (b) acetone, (c) butyraldehyde, and (d) butanone.
Caption: Figure 5: Full scan mass spectrum of tobacco smoke.
Table 1: Performance parameters of the APCI-MS/MS method for six low molecular carbonyl compounds by daughter scan mode with [[C[H.sub.3]O].sup.+]. Analytes Linear equation Linear range LOD ([micro]g/L) Acrolein y = 7.45e4x + 1.76e2 0.025~2.5 0.007 Acetone y = 1.95e5x + 2.3e3 0.05~5 0.021 Propionaldehyde y = 1.37e5x + 9.93e2 0.025~2.5 0.008 Crotonaldehyde y = 3.53e5x + 5.62e2 0.01~2.5 0.004 Butanone y = 2.13e5x + 1.3e3 0.05~5 0.012 Butyraldehyde y = 6.47e4x + 7.54e2 0.025~2.5 0.006 Acetone y = 1.25e5x + 9.88e2 0.05~5 0.009 Propionaldehyde y = 1.02e6x + 1.58e3 0.01~2.5 0.003 Butanone y = 3.26e4x + 4.71e2 0.05~5 0.013 Butyraldehyde y = 1.21e5x + 4.06e1 0.025~2.5 0.003 Analytes [R.sup.2] CID energy Acrolein 0.9979 15 eV Acetone 0.9973 15 eV Propionaldehyde 0.9993 15 eV Crotonaldehyde 0.9998 15 eV Butanone 0.9997 15 eV Butyraldehyde 0.9976 15 eV Acetone 0.9996 5 eV Propionaldehyde 0.9993 5 eV Butanone 0.9986 5 eV Butyraldehyde 0.9995 5 eV
|Printer friendly Cite/link Email Feedback|
|Title Annotation:||Research Article|
|Author:||Zhao, Wuduo; Zhang, Qidong; Lu, Binbin; Sun, Shihao; Zhang, Shusheng; Zhang, Jianxun|
|Publication:||Journal of Analytical Methods in Chemistry|
|Date:||Jan 1, 2017|
|Previous Article:||Separation and Determination of Fe(III) and Fe(II) in Natural and Waste Waters Using Silica Gel Sequentially Modified with Polyhexamethylene...|
|Next Article:||Simultaneous Determination of 11 Compounds in Gualou Guizhi Granule and Pharmacokinetics Study by UPLC-MS/MS.|