Excimer laser assisted implantable cardioverter defibrillator lead extraction: an alternative treatment to the surgery?/Lazer yardimiyla yerlestirilebilen kardiyoversiyon defibrilator lead cikarilmasi: cerrahi tedaviye bir secenek olabilir mi?
Different methods have been used for the lead extraction. Open chest surgery, manual traction, extend weight assisted traction, forceps assisted traction and mechanical extraction systems with locking stylet have been used (2). Manual traction is often effective to remove leads implanted no more than a few months. However, chronically implanted leads require more advanced extraction systems (1,3).
Successful, economical, safe and predictable methods would also justify the removal of redundant leads presently abandoned. To address this problem an excimer laser sheaths was developed by the Spectranetics Corporation (Colorado Springs, Colorado) to be incorporated into the system of traction-counter traction for removal of chronically implanted leads (4).
In this article, we retrospectively reviewed and analysed our experience forthe excimer laser assisted lead extraction in patients with ICD. Local anesthesia associated with sedation was used in all patients, interventions were performed in the catheterization laboratory. Surgeons were in stand-by for any complication in the operating room of cardiovascular surgery. Any simple traction tried and the procedures were primarily planned for laser extraction. Implantation durations were very long. For this reason we used excimer laser assisted lead extraction system. After opening the ICD pocket, the pulse generator was removed and subcutaneous tissue surrounding the lead is dissected. A locking stylet is inserted into the inner coil and locked with leading end. The laser sheath was passed over the locking stylet and the lead. A tightly fitting polypropylene sheath was mounted on the outside of the laser sheath to disrupt fibrotic bands. The laser sheath was then advanced along the lead to venous entry. Lasing was started and two sheaths (laser and non-laser) advanced with fluoroscopic guiding. The non-laser outer sheath was advanced with laser sheaths for mechanical ablation when necessary. Laser lights were applied for 5 seconds intermittently. Freeing adherences of the lead to the venous and cardiac structures until the distal portion of lead was reached by the laser sheath. At this time laser light was interrupted, laser sheath was used for supporting the myocardium and applying traction to the lead. The tip of the lead was carefully extracted from myocardium.
Four consecutive patients underwent ICD lead extraction. The mean age was 44[+ or -]7 years, 3 were male and the mean duration of implantation was 42[+ or -]15 months. Underlying heart diseases were ischemic cardiomyopathy in 2 patients, arrhythmogenic right ventricular dysplasia in 1 patient, and sudden cardiac death in 1 patient. All types of arrhythmia were ventricular tachycardia and all devices were ICD. Indications for extraction were pocket infection in 2 patients, lead fracture in 1 patient, and lead dysfunction in 1 patient. Mean total extraction time was 39[+ or -]3 minutes. All of leads were extracted successfully and we did not experience any major or minor complication (Table 1, Fig. 1).
Infection is always the major indication for the lead extraction. In many patients, the lead infection can not be treated with antibiotics and the extraction of infected leads is always required. After implantation, ICD and pacemaker leads undergo a fibrotic encapsulating process by activation of humoral and cellular mechanisms. Once lead has been in place for many months, fibrous adhesions tend to increase at the contact points between lead and venous or cardiac walls. By the time these adhesions become dense and may extend along the length of the lead (1).
The use of excimer laser assisted lead extraction system was first reported in 1996 and success rate of this system was between 81%-100% (5,6). In 1997 Wilkoff et al. (7) compared excimer laser assisted lead extraction system with teflon telescoping sheaths in the "Pacing Lead Extraction with the Excimer Sheath" (PLEXES) trial with 301 patient. Complete lead removal rate was 94% in the excimer laser group and 64% in the other group (p=0.001). Partial extraction was 2.5% and failure was 3.3% in the laser group. The use of excimer laser reduced the duration of the intervention compared to non-laser group. After PLEXES trial, the US Laser Sheath Registry reported their experience in 863 patients with 1285 leads at 52 centers. The clinical success rate was 92% with excimer laser system (8).
[FIGURE 1 OMITTED]
Kennergen et al. (9) reported the results of European Multicenter study in 149 patients. Complete extraction rate was 89.5% and partial extraction rate was 6%. Roux et al. (10) also reported their series with 175 patients in 2007. Between September 2000 and August 2005, a total of 270 leads were extracted with excimer laser sheath system. 241 leads (89 %) were completely, 7 leads (3%) partially extracted. These findings supports that the excimer laser assisted lead extraction system is highly effective.
Major complications of excimer laser assisted lead extraction system are cardiac tamponade, tricuspid laceration, pneumothorax, subclavian vein laceration with massive hemothorax, and transient blood pressure drop prompting surgical exploration (7,8). Minor complications are local bleeding, small hematoma, small pericardial effusion, pulmonary edema and transient low output state (9,10). In PLEXES trial, complication rate was 2.6%. One death and three major complications were observed (7). Complication rate was 2.1% in the US registry, 2.7% in European Multicenter Study and 3.4% in Roux's report (8-10). The most important limitation of this system is the presence of dense adherences containing calcium. On fluoroscopy it is not always possible to identify the presence of calcium (3). Upsizing of the sheath to go over the calcified area allowed laser to continue (8).
We have demonstrated that excimer laser assisted lead extraction system is efficient without any complication. The excimer laser assisted system allowed successfully extraction of the all ICD leads. This technique can be used for the vast majority of lead extraction as an alternative treatment to the surgery, but it should be left for experienced centers with cardiac surgery in stand-by and open chest surgery is still required for special cases and for complications. This technique is not a procedure to be performed in any small hospital implanting pacemakers.
(1.) Love CJ, Wilkoff BL, Byrd CL, Belott PH, Brinker JA, Fearnot NE, et al. Recommendations for extraction of chronically implanted transvenous pacing and defibrillator leads: indications, facilities, training. North American Society of Pacing and Electrophysiology Lead Extraction Conference Faculty. Pacing Clin Electrophysiol 2000; 23: 544-51.
(2.) Smith HJ, Fearnot NE, Byrd CL, Wilkoff BL, Love CJ, Sellers TD. Five-years experience with intravascular lead extraction. U.S. Lead Extraction Database. Pacing Clin Electrophysiol 1994; 17: 2016-20.
(3.) Gilligan DM, Dan D. Excimer laser for pacemaker and defibrillator lead extraction: techniques and clinical results. Lasers Med Sci 2001; 16:113-21.
(4.) Byrd CL. Extracting chronically implanted pacemaker leads using the Spectranetics Excimer Laser: Initial clinical experience. Pacing Clin Electrophysiol 1996; 19: 567-7.
(5.) Kennergren C. First European experience using excimer laser for the extraction of permanent pacemaker leads. Pacing Clin Electrophysiol 1998; 21:268-70.
(6.) Krishnan SC, Epstein LM. Initial experience with a laser sheath to extract chronic transvenous implantable cardioverter-defibrillator leads. Am J Cardiol 1998; 82:1293-5.
(7.) Wilkoff BL, Byrd CL, Love CJ, Hayes DL, Sellers TD, Schaerf R, et al. Pacemaker lead extraction with the laser sheath: results of the pacing lead extraction with the excimer sheath (PLEXES) trial. J Am Coll Cardiol 1999; 33:1671-6.
(8.) Epstein LM, Byrd CL, Wilkoff BL, Love CJ, Sellers TD, Hayes DL, et al. Initial experience with larger laser sheaths for the removal of transvenous pacemaker and implantable defibrillator leads. Circulation 1999; 100: 516-25.
(9.) Kennergren C. Excimer laser assisted extraction of permanent pacemaker and ICD leads: present experiences of a European multi-centre study. Eur J Cardiothorac Surg 1999; 15: 856-60.
(10.) Roux JE Page P, Dubuc M, Thibault B, Guerra PG, Made L, et al. Laser lead extraction: predictors of success and complications. Pacing Clin Electrophysiol 2007; 30: 214-20.
Hasan Gungor, Hamza Duygu, Bekir Serhat Yildiz, Ilker Gul, Mehdi Zoghi, Mustafa Akin
Department of Cardiology; Faculty of Medicine, FIge University, Izmir, Turkey
Address for Correspondence/Yazisma Adresi: Dr. Hamza Duygu, Ege University Medical School, Department of Cardiology, Izmir, Turkey Phone: +90 232 390 40 01 Fax: +90 232 390 32 87 E-mail: firstname.lastname@example.org
Table 1. Patient's demographics, extraction indications and results Underlying Size of Case Age, Type of heart locking number years Gender arrhythmia disease stylet 1 39 M VT ARVD LLD #2 2 37 F VT Sudden LLD #2 cardiac death 3 49 M VT Ischemic LLD #1 CMP 4 52 M VT Ischemic LLD #2 CMP Size of Implantation Case laser, duration. Indication for Complete number sheath months extraction removal 1 14F 51 Pocket Yes infection 2 14F 58 Lead Yes dysfunction 3 12 F 25 Lead Yes fracture 4 14F 34 Pocket Yes infection Total Case extraction number time, minutes Complication 1 43 No 2 38 No 3 42 No 4 36 No ARVD- arrhythmogenic right ventricular dysplasia, CMP- cardiomyopathy, F- female, LLD- lead locking device, M- male, VT- ventricular tachycardia
|Printer friendly Cite/link Email Feedback|
|Title Annotation:||Scientific Letter/Bilimsel Mektup|
|Author:||Gungor, Hasan; Duygu, Hamza; Yildiz, Bekir Serhat; Gul, Ilker; Zoghi, Mehdi; Akin, Mustafa|
|Publication:||The Anatolian Journal of Cardiology (Anadolu Kardiyoloji Dergisi)|
|Date:||Aug 1, 2009|
|Previous Article:||Differential diagnostic algorithm for diseases manifested with heart murmurs syndrome/Kalp ufurumu sendromu ile beraber gorulen hastaliklarda ayirici...|
|Next Article:||Assessment and follow-up of coronary abnormalities in Turkish children with Kawasaki disease/ Kawasaki hastaligi olan Turk cocuklarinda koroner arter...|