Barbed suture, now in the toolbox of minimally invasive gyn surgery: this recent addition to the laparoscopy tray was developed to resist migration and to allow you to approximate tissue more securely and dispense with knots.
* the Quill bidirectional barbed suture (Angiotech Pharmaceuticals) was approved by the US Food and Drug Administration (FDA) in 2004 for soft-tissue approximation and has been sold in the United States since 2007 (FIGURE 1A, page 40) (1)
* the V-Loc 180 unidirectional barbed suture (Covidien) won FDA approval in March for soft-tissue approximation. It should be available in October (FIGURE 1B).
(Note: We have performed laparoscopic surgery using the Quill suture, but not the V-Loc 180, both as part of clinical research and in practice.)
In the bidirectional suture, barbs are cut into suture in a direction opposite to the needle; the barbs change direction at the midpoint of the suture, and needles are swaged onto both ends (FIGURE 1A). (2)
The anchoring of bidirectional barbed suture resists migration and can be conceptualized as a "continuous interrupted" suture without knots; it has been shown to have tissue-holding performance that is at least equal to knot-anchored suture. (3,4)
A suture with bidirectional barbs offers several advantages:
* Because it self-anchors and is balanced by the countervailing barbs, no knots are required
* It self-anchors every 1 mm of tissue, yielding more consistent wound opposition; this may result in a more "watertight" seal
* Because it is knotless, it can securely reapproximate tissues in less time, at less cost, and with less aggravation. (5,6)
[FIGURE 1 OMITTED]
Note: Whether these characteristics apply to unidirectional barbed suture remains to be determined.
Because the effective diameter of suture is decreased when barbs are cut into it, barbed suture is rated one suture size smaller than smooth suture. A 0 barbed suture, for example, equals a 2-0 smooth suture.
In the current version of the Quill suture, there is a segment of approximately 7 cm of smooth suture proximal to each needle. A second-generation Quill will be released this year, with barbs extending all the way to the needles. Bidirectional barbed suture materials currently include polydioxanone (PDO), poliglecaprone 25 (Monoderm), nylon, and polypropylene.
The V-Loc 180 is a unidirectional barbed polyglyconate (Maxon) suture with a loop at the distal end to facilitate initial fastening.
We have performed more than 200 laparoscopic surgeries using Quill suture since March 2008; most involved total laparoscopic hysterectomy and laparoscopic myomectomy. We have also used Quill suture for laparoscopic sacrocolpopexy and uterosacral ligament suspension.
We have had outstanding clinical results: no cuff evisceration and only minor cases of vaginal bleeding or spotting in the immediate postoperative period. (7) In one case, reoperation was necessary after laparoscopic myomectomy, but the repeat surgery was performed to treat small-bowel obstruction that was not directly related to the hysterotomy repair.
For vaginal-cuff closure, we use 0 PDO suture on a 36-mm half-circle taper-point needle. We initially used half of a 14 x 14-cm suture with a LapraTy clip on the distal end. We began the repair at the distal end of the vaginal cuff, taking care to incorporate the uterosacral ligament into the initial bite, and continuing proximally until the other uterosacral ligament was incorporated into the repair.
More recently, we have been using the 7 x 7-cm 0 PDO because it is now available with the 36-mm half-circle taper-point needle. Regardless of suture material, it is important to obtain a full-thickness bite, with a 1-cm margin on the vaginal mucosa on each bite.
With the shorter suture, we begin closure in the middle of the cuff and take each needle to the opposite end of the cuff (FIGURE 2). We incorporate the uterosacral ligament into the final bite on each side, and we cut the suture without a knot or a LapraTy clip.
[FIGURE 2 OMITTED]
[FIGURE 3 OMITTED]
For myomectomy closure, we use 14 x 14-cm 0 PDO suture on a 36-mm half-circle taper-point needle. If the hysterotomy is longer than 8 cm, we prefer to use 24 x 24-cm suture. We tack the first needle to the opposite anterior abdominal wall to avoid having the suture become entangled. We close the deepest layer using the first needle; we use the second needle to close the more superficial layer and the serosa, if possible (FIGURE 3). If the suture is used beyond the barbed portion, we cut the needles and apply a LapraTy clip. Sometimes, three or four layers are needed to close a deep myometrial defect. 2/0 Monoderm can also be used for the serosa, continuously or as a baseball stitch.
Because bidirectional barbed suture is knotless, it can securely re-approximate tissues in less time than traditional suturing and at less expense
In myomectomy closure, use the first barbed suture to close the deepest myometrial layer
(1.) Department of Health and Human Services letter approving Seclion 510(k) premarket notification for Quill suture. Available at www.accessdata.fda.gov/cdrhdocs/ pdf4/k042075.pdf. Accessed August 13, 2009.
(2.) Leung JC. Barbed suture technology: recent advances. In: Medical Textiles 2004, Conference Proceedings; Oct. 26-27, 2004; pp. 62-80; Pittsburgh.
(3.) Bashid RM, Sartori M, White LE, Villa MT, Yoo gg, Alam M. Breaking strength of barbed polypropylene sutures: rater-blinded, controlled comparison with nonbarbed sutures of various calibers. Arch Dermatol. 2007;143:869 872.
(4.) Rodeheaver GT, Pifieros-Fernandez A, Salopek LS, et al. Barbed sutures for wound closure: in vivo wound security, tissue compatibility and cosmesis measurements. In: Society for Biomaterials 30th Annual Meeting Transactions; 2004; transaction 229, p. 232.
(5.) Greenberg JA, Einarsson JI. Use of bidirectional barbed suture in laparoscopic myomectomy and totallaparoscopic hysterectomy. J Minim Invasive Gynecol. 2008;15:621-623.
(6.) Moran ME, Marsh C, Perrotti M. Bidirectional-barbed sutured knotless running anastonrosis v classic Van Velthoven suturing in a model system. J Endourol. 2007;21:1175-1178.
(7.) Einarsson JI, Vellinga T, Twijnstra A, Suzuki Y, Greenberg JA. The use of bidirectional barbed suture in laparoscopic mymnectonry and total laparoscopic hysterectomy; an evaluation of safety and clinical outcomes. Submitted as an abstract to the 38th Global Congress in Minimally Invasive Gynecology, 2009.
Jon I. Einarsson, MD, MPH
Dr. Einarsson is Director of Minimally Invasive Gynecologic Surgery at Brigham and Women's Hospital and Assistant Professor of Obstetrics, Gynecology, and Reproductive Biology at Harvard Medical School in Boston.
James A. Greenberg, MD
Dr. Greenberg is Chief of Gynecology at Faulkner Hospital and Vice Chair of the Department of Obstetrics and Gynecology at Brigham and Women's Hospital in Boston.
The authors report no financial relationships relevant to this article.
|Printer friendly Cite/link Email Feedback|
|Title Annotation:||SURGICAL TECHNIQUES|
|Author:||Einarsson, Jon I.; Greenberg, James A.|
|Date:||Sep 1, 2009|
|Previous Article:||What's the best approach to managing chronic pain? Take an interdisciplinary approach and avoid making medication the sole focus of treatment,...|
|Next Article:||Vessel--sealing devices: this overview of vessel-sealing devices for laparoscopic procedures should help you select a system that meets the...|