Analisis de concordancia de diferentes metodologias para la identificacion de aislamientos orales de especies de Candida.
The genus Candida consists of a group of ubiquitous yeasts with diverse characteristics. The best-known species in the group is Candida albicans, as it is the main species related to most yeast infections in humans, however, an increase in the number of infections caused by species different from C. albicans has been observed. These emerging species have become more important, since some have profiles of resistance to commonly used antifungal drugs, especially azoles and echinocandins (1). Based on the above, the correct identification of yeasts of the genus Candida is one of the greatest challenges at present, because this could delay the establishment of suitable treatment in patients with invasive fungal infections. For this reason, it is necessary to have fast and accurate tests for identification of clinical isolates (2).
Currently, there are several methodologies for the identification of yeasts, some using chromogenic medias such as CHROMagar[TM] Candida, that allow a presumptive identification of the most common and relevant clinical species (C. albicans and C. tropicalis). It is important to know that many authors emphasize the importance of complementing such identification with other phenotypic tests that allow species confirmation (3). Similarly, there are commercial systems such as API[R] 20 C AUX or Vitek[R] 2 Compact system for the identification of yeasts, using methodologies that are based on biochemical identification. They have the disadvantage that they can generate misidentification due to the lack of experience of the laboratory technician in the interpretation of results, and with some frequency these systems are not able to differentiate species with similar biochemical profiles, because some emergent species are not included in their databases (4). Mass spectrometry, based on the matrix-assisted laser desorption ionization time-of-flight (MALDI TOF) methodology, has emerged as a valuable method for the identification of microorganisms, and with good performance in the identification of yeasts, for its speed and precision (5,6). For these reasons, this technology has been used more frequently in clinical laboratories, the commercial systems Microflex' (Bruker Daltonics GmbH, Leipzig, Germany) and Vitek[R] MS (bioMerieux, Marcy, L'Etoile, France) being the most popular (6,7).
Other methodologies employ molecular techniques based on the sequencing of nucleic acids, which has been used as a reference method to make comparisons with other identification tests, because it provides more accurate identification (8,9). Additionally, sequencing techniques permit the identification of cryptic species such as C. orthopsilosis, C. nivariensis, or C. bracarensis with high precision, which frequently exhibit resistance to antifungal agents 9. As a limitation, the use of DNA sequencing techniques is limited to laboratories with special spaces, has equipment requirements, and requires highly trained personnel (10).
Considering that there are few studies that have analyzed the concordance of yeasts identification methods of, the principal aim of this study was to evaluate the concordance between five different methods, based on biochemical testing, mass spectrometry and DNA sequencing, that are used for the identification of yeasts of the genus Candida.
Materials and Methods
Population and study site
Oral rinses were obtained during the 2014 from 98 healthy adult individuals attending at the Universidad Antonio Narino dental clinics, located in nine Colombian cities (Armenia, Bogota, Bucaramanga, Cucuta, Ibague, Neiva, Palmira, Popayan and Villavicencio). These individuals did not have known systemic disease, although some had localized pathological processes. Samples from patients with systemic disease and from patients receiving antibiotic, antifungal or corticosteroid treatment in the last 6 months were excluded from the analyses.
Mouthwashes were immediately submitted to the Laboratory of the Medical and Experimental Mycology Unit, at the Corporacion para Investigaciones Biologicas (CIB) in Medellin, Colombia. Samples were processed using Sabouraud Dextrosa[TM] agar with antibiotics (BD[TM], reference 210950) 11 and with CHROMagar[TM] Candida (CHOMagar Microbiology, Paris, France) (12), which allowed verification of the possibility of mixed infections by several species of Candida from the primary cultures. The cultures were incubated at 25[degrees]C for 20 days, with weekly readings to evaluate the type of growth. The recovered isolates (n= 67) were stored in sterile distilled water at 4[degrees]C and in a medium composed of skim milk (BD[TM], reference 232100) at -20[degrees]C.
Identification of yeast
Identification of the isolates was performed using the following methodologies: 1) CHROMagar[TM] Candida (CHOMagar Microbiology, Paris, France). 2) API[R] 20 C AUX (bioMerieux, Marcy, L'Etoile, France). 3) Vitek[R] 2 Compact automated system (bioMerieux, Inc., Hazelwood, MO, USA). 4 and 5) Mass spectrometry based on the MALDI TOF MS (matrix assisted laser desorption / ionization) technique in Vitek[R] MS (bioMerieux, Marcy, L'Etoile, France) and Microflex' (Bruker Daltonics GmbH, Leipzig, Germany). 6) Polymerase Chain Reaction (PCR) Panfungal and sequencing.
Methodologies used for the identification of yeasts
1. CHROMagar[TM] Candida: After surface growth on this medium, the color of each of the colonies was checked to classify them according to the following characteristics: C. albicans/ dubliniensis complex colonies showed medium to light green color, C. tropicalis blue colonies, and other species presented pink or light to dark lilac color, or their natural cream color (12,13).
2. API[R] 20 C AUX: Testing was performed following the manufacturer's recommendations (bioMerieux, Marcy-l'Etoile, France) 14. After the incubation period (48 h at 25[degrees]C), panels were checked visually. The numerical profile obtained for each isolate was interpreted using the ApiwebTM software (bioMerieux, reference: 40 011).
3. Vitek[R] 2 Compact: The inoculum was prepared in 3 mL of 0.45% saline solution, using a pure culture of no more than 24 h growth. The suspension was adjusted to a McFarland turbidity range of 1.8-2.2 using the DensiCheck[R]. The final inoculum was automatically dispensed into the kit identification cards (YST, reference: 21343), and incubated using the Vitek[R] 2 Compact equipment (bioMerieux, Durham, NC). The final identification was classified as follows: "excellent," "very good" "good," "acceptable" or "with low discrimination" depending on confidence level and percentage of discrimination for each identification provided by the equipments software (15).
4. MALDI TOF MS: This was done using two commercial platforms; the Vitek[R] MS equipment (bioMerieux, Marcy, L'Etoile, France) and the Microflex' (Bruker Daltonics GmbH, Leipzig, Germany) Biotype Library 3.0. The methodological details corresponding to each commercial method are described below.
4.1 Vitek[R] MS: A single pure colony (growth 24-72 h) was deposited in a single well of the Vitek[R] MS plate. Cells were lysed with 0.5 [micro]L of 25% concentration formic acid (Ref: 411072) and allowed to dry at room temperature (1-2 min). After drying, 1 [micro]L of the CHCA matrix (bioMerieux, Marcy, L'Etoile, France, reference: 411071) was added. Tests were performed after the final mixture was completely dry. The peak spectrum was analyzed using the MS-ID server (MS-ID [CE/IVD] database) (16).
4.2 Microflex': Isolates grown for no more than 24 h at 37[degrees]C were tested by direct extraction methodology in plate or by extraction with formic acid. For direct extraction, a single colony was deposited directly in the MALDI TOF plate, allowed to dry at room temperature; after dry, 1 pL of 100% concentration formic acid was added, and then was covered with 1 [micro]l of HCAA matrix ([alpha]-cyano-4 hydroxycinnamic acid - HCCA). After drying for a second time at room temperature, the treated samples were analyzed using the Biotyper machine and the final mass spectra were analyzed using the FlexControl software (version 3.0) and the MALDI Biotyper RTC. Strains that had a low score in the direct test identification or in which the identification was not possible were extracted with formic acid, following the manufacturer's instructions (Bruker Daltonik GmbH, Bremen, Germany) (17). Each sample was served in duplicate to verify the reproducibility.
5. Panfungal PCR and sequencing: The D1/D2 region of the 28S rRNA gene was amplified, following the international guidelines for molecular identification of fungi (18). The ITS 1-4 (Internal Transcribed Spacer) region was also amplified for the identification of cryptic species in Candida parapsilosis and Candida glabrata isolates (13). Genomic DNA was extracted from isolated colonies grown on Sabouraud agar using the QIAamp DNA mini kit (QIAGEN, Germantown, MD), following the manufacturer's recommendations. Molecular markers were amplified using primers and protocols previously described for the D1/D2 and ITS 1-4 regions (19,20). Amplified products from the D1/D2 region (~600 bp) and the ITS region 1-4 (600-900 bp) were shipped to Macrogen laboratories (Maryland, USA) for sequencing. Editing and aligning of sequences were performed using the Sequencher 5.0 software (Gene Code Corporation). A search was made to establish similarity/homology in two databases: the NCBI (BLAST) (National Biotechnology Information Center, Washington, DC) and the CBS-KNAW (Fungal Diversity Center).
Methodological design and statistical analysis
Variables analyzed in this study were summarized by calculating both absolute and relative frequencies. A concordance analysis was performed to evaluate the agreement between different methods used for yeast identification. The Weighted Kappa (wK) values and their respective 95% confidence intervals (95% CI) were calculated. The concordance analysis was performed in two ways, as follows: The first analysis was done by regrouping the results obtained by the API[R] 20 C AUX, Vitek[R] 2 Compact, Vitek[R] MS, Microflex' and panfungal PCR and sequencing, in the following three categories of results: C. albicans/dubliniensis, C. tropicalis, and Candida spp different than Candida albicans/dubliniensis/ tropicalis. This regrouped result was done to compare the five previous mentioned methodologies against the CHROMagar[TM] Candida. The second analysis compared the results of the API[R] 20 C AUX, Vitek[R] 2 Compact, Vitek[R] MS, Microflex[R] and panfungal PCR and sequencing, and in this second analysis the panfungal PCR and sequencing method was selected as the reference methodology. In addition, a cost/turnaround time results analysis was performed, using the sale price of the service and the turnaround time (excluding the times of the pre-analytical and post-analytical phases) from local laboratory providers in the city of Medellin, Colombia. Information was collected through telephone calls or email. Interpretation was performed using graphic analysis of the two variables (cost and turnaround time). Statistical analyses were performed using the statistical package STATA 8.0[R] and graphics using Microsoft Excel 2010[R] software.
The protocol was approved by the ethical committees of the Universidad Antonio Narino, Armenia - Quindio, Colombia.
Sixty-seven Candida isolates were recovered from 98 oral rinses (68% positiviy). In the initial analysis using the CHROMagar[TM] Candida the 67 isolates were classified as follows: 39 (58%) isolates as C. albicans/dubliniensis, 4 (6%) as C. tropicalis and 24 (36%) isolates such as Candida different than Candida albicans/ dubliniensis/tropicalis. The first concordance analysis to compare the CHROMagar[TM] Candidaagainst the other five methodologies (regrouped results) gave the following Weighted Kappa (wK) results: vs API[R] 20 C AUX = 1.00 (95% CI= 1.00-1.00); vs Vitek[R] 2 Compact = 0.87 (95% CI= 0.75-0.96); vs Vitek[R] MS= 0.92 (95% CI= 0.80-0.99); vs Microflex[R] = 0.97 (95% CI= 0.94-1.00); and vs panfungal PCR and sequencing = 0.98 (95% CI= 0.95-1.00).
In the results of yeast identification, excluding the CHROMagar[TM] Candida results, it was observed that Panfungal PCR and sequencing was the method that identified the largest number of species (n= 12), following by Mass spectrometry methods, Vitek[R] MS and Microflex[R], which identified 9 different Candida species each, and finally the biochemical methods API[R] 20 C AUX and Vitek[R] 2 Compact, which identified 8 different Candida species each. The description of the species identified by each methodology is summarized in Figure 1.
The second concordance study showed a high concordance between the results obtained with Panfungal PCR and sequencing, Vitek[R] MS, Microflex[R] and API[R] 20 C AUX, these results were grouped into the good and very good categories. The wK that involved the Vitek[R] 2 Compact method were characterized by moderate or good results compared with the other methods (Panfungal PCR and sequencing, Vitek[R] MS, Microflex[R] and API[R] 20 C AUX). A detailed analysis of the concordances and discordances comparing all the methods is given in Table 1. Values of wK observed and their respective 95% CI are summarized in Figure 2.
To confirm that the previous methods did not fail to detect species of the complex Candida parapsilosis and Candida glabrata, a PCR and sequencing analysis of the ITS 1-4 region was performed, determining no presence of cryptic species.
The analysis of costs/turnaround time results showed that the methods based on the MALDI TOF technology (Microflex[R] and Vitek[R] MS) required less time to generate a result (4 minutes), and in addition, the Microflex[R] method had the lower commercial cost (13 US Dollars). The panfungal PCR and sequencing was the method that required most time to generate a final report (3 days) and had the highest commercial value (86 USD). The analysis that compares the costs and turnaround time results for the tests evaluated in this report is summarized in Figure 3.
This study evaluated five different methodologies for the identification of clinical isolates of the Candida genus obtained from oral rinses. In general, we observed good concordance between the CHROMagar[TM] Candida and the other five methodologies evaluated in this study. The Vitek[R] 2 Compact method had the lowest agreement and the highest agreement was observed with the API[R] 20 C AUX. It is important to mention that use of CHROMagar[TM] Candida allows for identification of the presence of multiple species in the same clinical sample and gives a presumptive identification of the associated Candida species (21,22).
Most of the five methods analyzed showed high concordances grouped into good and very good categories. Previous studies have shown that phenotypic methods, based on biochemical tests such as the API[R] 20 C AUX and the Vitek[R] 2 Compact system, are the most used in clinical laboratories (23,24). In this study, it was observed that these two methods had the lowest capacity to differentiate Candida species and also presented lower values of wK, especially when comparing the Vitek[R] 2 Compact method with the other methodologies. Other authors have reported that these methods may present discrepancies in identifying less common Candida species and especially those that are closely related (23,25). In addition, the accuracy of these methods is greatly reduced if the laboratory technician does not perform the additional tests required for the identification.
In our study, Panfungal PCR and sequencing was able to identify species such as Candida fermentati and Candida haemulonii, which are emerging species, recently classified as species complex, that are difficult to identify by conventional methods and whose identification can become a challenge, because these species have not been included in the databases of most of the different identification systems available (26,27). In addition, some of these emerging species have shown a decrease in sensitivity to antifungal agents, and consequently, this increases the difficulty in the management of patients with invasive Candida infections (26-28). In our study, none of the methodologies analyzed, except for Panfungal PCR and sequencing, had the ability to identify Candida lipolytica (n= 1), Candida fermentati (n= 1) and Pichia kluyveri (n= 1), species that have been reported mainly for industrial use, but, also have the capacity to behave as pathogens in humans according to host conditions (26,29-32).
The methodologies based on MALDI TOF (Microflex[R] and Vitek[R] MS) presented high concordance compared with the reference method, panfungal PCR and sequencing. The analysis of concordance between these two technologies showed a very good agreement, due to the fact that both systems identified the same number of species, but not all of them corresponded to the same species, since there were inconsistencies in the identifications in 6 of the 67 analyzed isolates. There are several platforms of MALDI TOF, and differences will depend on the libraries of mass spectra and the algorithms of identification that each platform has (33,34). In this study the Microflex[R] was able to identify all the isolates at species level. When using the Vitek[R] MS spectrometer, three major discordances were observed (this platform was not able to determine the final identification at species level in three isolates), when compared with the reference method, affecting its wK value. Many studies have shown that these techniques are faster and more accurate in the identification of yeasts from crops, which has a high relevance in terms of time and costs (7,35,36).
The panfungal PCR and sequencing advantage is that this technique allows the identification of cryptic species or complex species, but it is recommended to combine the amplification of the D1/D2 region with other targets in the ITS region (24,26). In this report, although the presence of cryptic species of Candida parapsilosis and Candida glabrata were not observed, it is important to use methodologies that are able to detect them, mainly due to the emergence of species that are difficult to manage (9,37,38). In addition, this technique would be the best option, despite its cost, for those identifications that are doubtful or where other methodologies have low discriminatory power.
When we analyzed the costs and turnaround time results, we could determine that the methods based on the MALDI TOF technology (Microflex[R] and Vitek[R] MS) were the ones that offered the least time for results, as previously reported, and when comparing this methodology with other conventional methods, our results also reflected a reasonable and competitive cost (39). It should be noted that this technology allows precise results, surpassing the conventional identification techniques (40,41). This technology has shown great potential for fast and accurate identification, and this finding demonstrates the need to make its methodologies more available.
It is important to mention that delays can occur in the generation of results, as a result of factors that are not inherent to the methodology or external, such as the day that the isolation arrives in the laboratory, the viability of the isolate, the need for additional extraction steps, the days designated in the laboratory for sample processing, the availability of laboratory personnel, or turnaround time for sequencing analysis. Changes in some of these factors can make a significant difference when delivering a timely outcome and it would help to implement changes at the laboratory level to improve the different processes.
The present study was limited to the evaluation of Candida species isolated from oral mucosa. Although it was possible to show a wide variety of species identified between the different methods, it would be pertinent for future investigations to carry out the same evaluation in other clinical specimens. In addition, the cost and time analysis was only carried out in the city of Medellin, Colombia. In Colombia the number of laboratories that offer molecular methods or MALDI TOF technology is limited, which may increase the costs of these tests in a significant way, mainly due to the exclusivity of the service.
It is necessary to implement identification methods that facilitate access to fast and reliable results, but at the same time, help to optimize the economic resources once those are implemented in the daily routine. In this work, it was shown that the systems based on the MALDI TOF technology, despite being methodologies that initially required a substantial economic investment, proved to be fast, economical (commercial value of the test) and precise, which are presented as promising alternatives for the routine identification of yeasts of the Candida genus. For those laboratories not able to access molecular or mass spectrometry tests, the use of automated tests could be a good alternative if the laboratorian technician follows the specification provided by the manufacture, and pays special attention to those results that involve uncommon species or discrepant characteristics presented in the isolate (morphology, susceptibility patter and other additional taxonomy keys).
The members of the Medical and Experimental Mycology Unit, Corporacion para Investigaciones Biologicas (CIB) express their appreciation to the following persons and institutions: the Universidad Antonio Narino, Quindio, Colombia, the Research Unit in Proteomics and Human Mycosis, Infectious Diseases Group, Pontificia Universidad Javeriana, Bogota, Colombia, the SAS Reference Medical Laboratory, the Committee for the Development of Investigation (CODI), University of Antioquia, Medellin, Colombia, and to Mr Mark Mezzullo and Dr Angela Restrepo for editing of the manuscript.
Universidad Antonio Narino, Quindio, Colombia and Corporacion para Investigaciones Biologicas (CIB), Medellin, Colombia
Conflicts of interest:
All other authors not report potential conflicts of interest. The findings and conclusions in this report are those of the authors and do not necessarily represent the official position of the Centers for Disease Control and Prevention.
(1.) McCarty TP, Pappas PG. Invasive candidiasis. Infect Dis Clin North Am. 2016; 30(1):103-24. McCarty TP, Pappas PG. Invasive candidiasis. Infect Dis Clin North Am. 2016; 30(1): 103-24.
(2.) Albataineh MT, Sutton DA, Fothergill AW, Wiederhold NP. Update from the laboratory: clinical identification and susceptibility testing of fungi and trends in antifungal resistance. Infect Dis Clin North Am. 2016; 30 (1): 13-35.
(3.) Estrada-Barraza D, Davalos A, Flores-Padilla L, Mendoza-De Elias R, Sanchez-Vargas LO. Comparacion entre metodos convencionales, CHROMagarTM Candida y el metodo de la PCR para la identificacion de especies de Candida en aislamientos clinicos. Rev Iberoam Micol. 2011; 28 (1): 36-42.
(4.) Ochiuzzi ME, Cataldi S, Guelfand L, Maldonado I, Arechavala A, Red de Micologia CABA, Argentina. Evaluacion del sistema Vitek(r) 2 para la identificacion de las principales especies de levaduras del genero Candida. Rev Argent Microbiol. 2014; 46(2): 107-110.
(5.) Panda A, Ghosh AK, Mirdha BR, et al. MALDI TOF mass spectrometry for rapid identification of clinical fungal isolates based on ribosomal protein biomarkers. J Microbiol Methods. 2015; 109: 93-105.
(6.) Jordana-Lluch E, Martro Catala E, Ausina Ruiz V La espectrometria de masas en el laboratorio de microbiologia clinica. Enferm Infecc Microbiol Clin. 2012; 30 (10): 635-44.
(7.) Lima-Neto R, Santos C, Lima N, Sampaio P, Pais C, Neves RP. Application of MALDI TOF MS for requalification of a Candida clinical isolates culture collection. Braz J Microbiol. 2014; 45 (2): 515-22.
(8.) Kathuria S, Singh PK, Sharma C, et al. Multidrugresistant Candida auris misidentified as Candida haemulonii: characterization by Matrix-Assisted Laser Desorption Ionization-Time of Flight Mass Spectrometry and DNA sequencing and its antifungal susceptibility profile variability by Vitek(r) 2, CLSI Broth Microdilution, and Etest Method. J Clin Microbiol. 2015; 53(6): 1823-30.
(9.) Criseo G, Scordino F, Romeo O. Current methods for identifying clinically important cryptic Candida species. J Microbiol Methods. 2015; 111: 50-6.
(10.) Vijayakumar R, Giri S, Kindo AJ. Molecular species identification of Candida from blood samples of intensive care unit patients by Polymerase Chain Reaction--Restricted Fragment Length Polymorphism. J Lab Physicians. 2012; 4 (1): 1-4.
(11.) Becton, Dickinson and Company. Regulatory Documents. EP/ USP TSM Media--Test for Candida albican Sabouraud Dextrosa. Becton, Dickinson and Company. Switzerland Sarl. Available from: http://www.bd.com/europe/regulatory/documents. asp?i=506.
(12.) Becton, Dickinson and Company. BBL CHROMagar Candida Medium. Instrucciones de uso--medios en placa listos para usar. 2014. PA 257480.05. Available from: http://www.bd.com/resource. aspx?IDX=8810
(13.) Howell SA, Hazen KC, Brandt ME. Candida, Cryptococcus, and other yeast of medical importance. Manual of clinical microbiology. Eleven Edition. Washington, DC: ASM Press; 2015. p. 1984-2014.
(14.) bioMerieux. API 20 C AUX. Yeast Identification System. bioMerieux, Marcy-l'Etoile: France. Available from: http://www. biomerieux.com.co/diagnostico-clinico/apir. Cited: 01/07/2018.
(15.) bioMerieux. VITEK 2 Compact. Yeast Identification System. bioMerieux, Marcy-l'Etoile: France. Available from: http://www. biomerieux.com.co/diagnostico-clinico/vitekr-2-compact. Cited: 01/07/2018.
(16.) bioMerieux. VITEK MS. bioMerieux, Marcy-l'Etoile: France. Available from: http://www.biomerieux.com.co/diagnostico clinico/vitekr-ms. Cited: 01/07/2018.
(17.) Bruker. MALDI-TOF Microflex. Bruker Daltonik GmbH, Bremen, Germany. Available from: https://www.bruker.com/ products/mass-spectrometry-and-separations/maldi-toftof/ microflex/overview.html. Cited: 01/07/2018
(18.) Clinical and Laboratory Standards Institute. Interpretive criteria for identification of bacteria and fungi by DNA target sequencing: approved guideline. MM18-A. Clinical and Laboratory Standards Institute: Wayne, PA; 2008.
(19.) Voigt K, Cigelnik E, O'Donnell K. Phylogeny and PCR identification of clinically important zygomycetes based on Nuclear Ribosomal-DNA sequence data. J Clin Microbiol. 1999; 37(12): 3957-64.
(20.) White TM, Bruns T, Lee S, Taylor J. Amplification and direct sequencing of fungal ribosomal RNA for phylogenetics. In: Innis MA, Gelfand DH, Sninsky JJ, White TJ. (Eds.). PCR protocols: a guide to methods and applications. Academic Press: San Diego, CA; 1990. pp. 315-21.
(21.) Stefaniuk E, Baraniak A, Fortuna M, Hryniewick W. Usefulness of CHROMagar Candida medium, biochemical methods--API[R] ID32C and Vitek 2 Compact and two MALDI TOF MS systems for Candida spp identification. Pol J Microbiol. 2016; 65 (1): 111-4.
(22.) Balleste R, Arteta Z, Fernandez N, Mier C, Mousques N, Xavier B, et al. Evaluacion del medio cromogeno CHROMagar Candida TM para la identificacion de levaduras de interes medico. Rev Med Uruguay. 2005; 21: 186-93.
(23.) Linton CJ, Borman AM, Cheung G, Holmes AD, Szekely A, Palmer MD, et al. Molecular identification of unusual pathogenic yeast isolates by large ribosomal subunit gene sequencing: 2 years of experience at the UK Mycology Reference Laboratory. J Clin Microbiol. 2007; 45(4): 1152-8.
(24.) Meletiadis J, Arabatzis M, Bompola M, Tsiveriotis K, Hini S, Petinaki E, et al. Comparative evaluation of three commercial identification systems using common and rare bloodstream yeast isolates. J Clin Microbiol. 2011; 49(7): 2722-7.
(25.) Borman AM, Szekely A, Palmer MD, Elizabeth M, Jonson EM. Assessment of accuracy of identification of pathogenic yeasts in microbiology laboratories in the United Kingdom. J Clin Microbiol. 2012; 50(8): 2639-44.
(26.) Romi W, Keisam S, Ahmed G, Jeyaram K. Reliable differentiation of Meyerozyma guilliermondii from Meyerozyma caribbica by internal transcribed spacer restriction fingerprinting. BMC Microbiol. 2014; 14: 52.
(27.) Cendejas-Bueno E, Kolecka A, Alastruey-Izquierdo A, Theelen B, Groenewald M, Kostrzewa M, et al. Reclassification of the Candida haemulonii complex as Candida haemulonii (C. haemulonii group I), C. duobushaemulonii sp. nov.(C. haemulonii group II), and C. haemulonii var. vulnera var. nov.: three multiresistant human pathogenic yeasts. J Clin Microbiol. 2012; 50(11): 3641-51.
(28.) Arendrup MC, Boekhout T, Akova M, Meis JF, Cornely OA, Lortholary O, et al. ESCMID and ECMM joint clinical guidelines for the diagnosis and management of rare invasive yeast infections. Clin Microbiol Infect. 2014; 20(Suppl 3): 76-98.
(29.) Crolla A, Kennedy K J. Fed-batch production of citric acid by Candida lipolytica grown on n-paraffins. J Biotechnol. 2004; 110(1): 73-84.
(30.) Amaya-Delgado L, Herrera-Lopez EJ, Arrizon J, ArellanoPlaza M, Gschaedler A. Performance evaluation of Pichia kluyveri, Kluyveromyces marxianus and Saccharomyces cerevisiae in industrial tequila fermentation. World J Microbiol Biotechnol. 2013; 29(5): 875-81.
(31.) Lockhart SR, Messer SA, Pfaller MA, Diekema DJ. Identification and susceptibility profile of Candida fermentati from a worldwide collection of Candida guilliermondii clinical isolates. J Clin Microbiol. 2009; 47(1): 242-4.
(32.) Liu WC, Chan MC, Lin TY, Hsu CH, Chiu SK. Candida lipolytica candidemia as a rare infectious complication of acute pancreatitis: a case report and literature review. J Microbiol Immunol Infect. 2013; 46(5): 393-6.
(33.) Mancini N, De Carolis E, Infurnari L, Vella A, Clementia N, Vaccaro L, et al. Comparative evaluation of the Bruker Biotyper and Vitek[R] MS Matrix-Assisted Laser Desorption Ionization-Time of Flight (MALDI TOF) Mass Spectrometry Systems for identification of yeasts of medical importance. J Clin Microbiol. 2013; 51(7): 2453-57.
(34.) Lohmann C, Sabou M, Moussaoui W, Prevost G, Delarbre JM, Candolfi E, et al. Comparison between the Biflex III-Biotyper and theAxima-SARAMIS Systems for Yeast Identification by Matrix-Assisted Laser Desorption Ionization-Time of Flight Mass Spectrometry. J Clin Microbiol. 2013; 51(4): 1231-6.
(35.) Pinto A, Halliday C, Zahra M, van Hal S, Olma T, Maszewska K, et al. Matrix-assisted laser desorption ionization-time of flight mass spectrometry identification of yeasts is contingent on robust reference spectra. PLoS One. 2011; 6(10): e25712.
(36.) Spanu T, Posteraro B, Fiori B, D'Inzeo T, Campoli S, Ruggeri A, et al. Direct MALDI TOF mass spectrometry assay of blood culture broths for rapid identification of Candida species causing bloodstream infections: an observational study in two large microbiology laboratories. J Clin Microbiol. 2012; 50(1): 176-9.
(37.) Brandt ME, Lockhart SR. Recent taxonomic developments with Candida and other opportunistic yeasts. Curr Fungal Infect Rep. 2012; 6 (3):170-7.
(38.) Tay ST, Lotfalikhani A, Sabet NS, Ponnampalavanar S, Sulaiman S, Na SL, et al. Occurrence and characterization of Candida nivariensis from a culture collection of Candida glabrata clinical isolates in Malaysia. Mycopathologia. 2014;178(3): 307-14.
(39.) Tan KE, Ellis BC, Lee R, Stamper PD, Zhang SX, Carrolla KC. Prospective evaluation of a matrix-assisted laser desorption ionization-time of flight mass spectrometry system in a hospital clinical microbiology laboratory for identification of bacteria and yeasts: a bench-by-bench study for assessing the impact on time to identification and cost-effectiveness. J Clin Microbiol. 2012; 50(10): 3301-8.
(40.) Iriart X, Lavergne RA, Fillaux J, Valentin A, Magnaval JF, Berry A, e t al. Routine identification of medical fungi by the new Vitek[R] MS matrix-assisted laser desorption ionization-time of flight system with a new time-effective strategy. J Clin Microbiol. 2012; 50(6): 2107-10.
(41.) Arvanitis M, Anagnostou T, Fuchs BB, Caliendo AM, Mylonakis E. Molecular and nonmolecular diagnostic methods for invasive fungal infections. Clin Microbiol Rev. 2014; 27(3): 490-526.
Alejandra Zuluaga , Karen Arango-Bustamante , Diego H. Caceres , Zilpa A. Sanchez-Quitian , Veronica Velasquez , Beatriz L. Gomez [1,4], Claudia M. Parra-Giraldo , Natalia Maldonado , Luz E. Cano [1,6], Catalina de Bedout , Raul E. Rivera 
 Unidad de Micologia Medica y Experimental, Corporacion para Investigaciones Biologicas (CIB), Medellin, Colombia.
 ORISE Fellow with the Mycotic Diseases Branch, Centers for Disease Control and Prevention (CDC), Atlanta, USA.
 Unidad de Investigacion en Proteomica y Micosis Humanas, Grupo de Enfermedades Infecciosas, Departamento de Microbiologia, Facultad de Ciencias, Pontificia Universidad Javeriana, Bogota, Colombia.
 Escuela de Medicina y Ciencias de la Salud, Universidad del Rosario, Bogota, Colombia.
 Laboratorio Medico de Referencia S.A.S, Grupo GERMEN, Medellin, Colombia.
 Escuela de Microbiologia, Universidad de Antioquia, Medellin, Colombia.
 Programa de Odontologia, Universidad Antonio Narino, Armenia--Quindio, Colombia.
* These authors contributed equally in this work
Received: 24 February 2018
Revised: 05 February 2018
Accepted: 23 July 2018
Diego H. Caceres, BSc, MSc. ORISE Fellow with the Mycotic Diseases Branch. CDC, Atlanta, USA. Phone: (+1) 470 776 5078. E-mail: xju7@cdc. firstname.lastname@example.org.
Caption: Figure 1. Distribution of species classified according to the identification methodology
Caption: Figure 2. Concordance analysis of five methodologies for the identification of oral isolates of Candida species.
Caption: Figure 3. Results of the cost/turnaround time analysis.
Table 1. Comparison of concordances and discrepancies of the API' 20 C AUX, Vitek' 2 Compact, Vitek' MS and Microflex' against the reference method Panfungal PCR and sequencing. Reference Method API[R] 20 C AUX Vitek[R] Panfungal PCR C (n) D (n) C (n) and sequencing (n) C. albicans 35 33 2 34 C. parapsilosis 12 12 0 10 C. dubliniensis 6 2 4 6 C. glabrata 4 4 0 2 C. tropicalis 2 2 0 2 C. intermedia 2 0 2 0 C. guilliermondii 1 1 0 0 C. lusitaniae 1 0 1 1 C. haemulonii 1 0 1 1 C. lipolytica 1 0 1 0 C. fermentati 1 0 1 0 Pichia kluyveri 1 0 1 0 Total 67 54 13 56 2 Compact Vitek[R] MS Microflex[R] D (n) C (n) D (n) C (n) D (n) C. albicans 1 33 2 35 0 C. parapsilosis 2 12 0 12 0 C. dubliniensis 0 5 1 5 1 C. glabrata 2 4 0 3 1 C. tropicalis 0 2 0 2 0 C. intermedia 2 2 0 1 1 C. guilliermondii 1 1 0 1 0 C. lusitaniae 0 1 0 1 0 C. haemulonii 0 1 0 1 0 C. lipolytica 1 0 1 0 1 C. fermentati 1 0 1 0 1 Pichia kluyveri 1 0 1 0 1 Total 11 61 6 61 6 n: numero C: concordances, D: discrepancies
|Printer friendly Cite/link Email Feedback|
|Title Annotation:||Original Article|
|Author:||Zuluaga, Alejandra; Arango-Bustamante, Karen; Caceres, Diego H.; Sanchez-Quitian, Zilpa A.; Velasque|
|Date:||Jul 1, 2018|
|Previous Article:||Olvidados, pero no perdonados: programas de inmunizacion hacia el siglo XXI.|
|Next Article:||Adaptacion y validacion de la lista de verificacion del parto seguro de la OMS, para su uso en Colombia.|