'[Re.sup.3] -inforced' converters for maximized patient protection.
Generally, the patient environment is defined as the area in which the patient may access medical equipment or where he may be connected to medical equipment in the course of an examination. To avoid electrical shock, the medical safety standard demands galvanic isolation between the supply, the equipment casing, and the diagnostic tools. Two safety barriers are paramount the primary barrier within the power supply of the equipment and a secondary barrier that isolates sensors and electrodes that may be in direct contact with the patient. If the primary isolation fails for any reason, the secondary isolation will still protect the patient.
[FIGURE 1 OMITTED]
Figure 1 shows the isolation features of a mains powered diagnostic unit intended for direct patient contact in this case, an ECG unit. For the secondary isolation of the body electrodes, DC/DC converters are used which feature reinforced isolation for additional safety.
The isolation of DC/DC converters approved for medical applications depends not only on the level of the isolation voltage, but more so on the quality of isolation.
For industrial applications, a functional isolation usually suffices, whereby the primary and secondary windings can be overlapped and the windings are insulated just by the coating on the wires. This method of construction gives high conversion efficiency because of the close proximity of the windings to each other. Also, although the insulating wire enamel is very thin, it can withstand up to 4.0 kVDC as long as it is not damaged. However, damage during the manufacturing process or excessive stresses in the windings can reduce the dielectric strength over time and can eventually cause isolation failure in the equipment.
UL (Underwriter Laboratories Inc.; www. ul.com) has thus defined different "qualities" of isolation: basic, supplementary, and reinforced (Table 1). The isolation class not only demands additional physical insulating barriers between input and output in case the functional isolation fails but also provides minimum creepage and clearance distances depending on the working voltage. For a basic isolation DC/DC-converter with an input voltage of up to 75 V, a clearance of 0.7 mm is mandatory. For a similar converter with reinforced isolation, the clearance must be 2.4 mm (i.e., three times larger). The same. applies for the creepage distance, which needs to be 4.6 mm for a reinforced converter instead of only 1.3 mm for a basic isolated one.
[FIGURE 2 OMITTED]
Lower Efficiency with Conventional Reinforced Technology
To meet the isolation class criteria, the primary and secondary windings can no longer be overlapping and must be physically apart. The winding scheme determines the size of the transformer and also the efficiency, since the magnetic fields are now no longer optimally overlapped with the increased separation.
An example: The efficiency of a conventionally isolated converter with overlapping primary and secondary windings is approximately 85%, while a reinforced converter in conventional design achieves only 75%.
The difference in efficiency has a marked influence on the performance of the converter. The losses for a standard isolated 3 Watt converter are around 500 mW (3.0 W/0,85 _ 3.0 W = 0.529 W), while they are twice as high for a conventionally manufactured reinforced-converter with the same output power (3.0 W/0.75-3.0W = 1.0W). The higher internal heat dissipation reduces the maximum operating temperature from 85[degrees]C down to only 75[degrees]C.
Development engineers at RECOM (www. recom-power.com) in Austria and Taiwan were able to design a transformer that meets all the conflicting requirements for wide clearance and creep age distances, multiple layers of insulation and an s compact size (Figure 2). This new concept has been filed for a patent under the designation "Re(3)-inforced." DC/DC converters incorporating this design achieve a higher isolation, better efficiency, and more rated power than comparable standard products. Apart from the higher isolation of up to 10 kVDC, it was also possible to reduce the winding capacitance by a factor of three down to 20 pF. This leads to extremely low leakage currents, also a common requirement for medical applications. These converters are continuously short circuit and overload protected and can optionally be supplied with under voltage lockout and a remote on/off pin. They are certified to EN60601-1, CSA C22.2 601-1, and UL60601-1.
Ask the Experts!
Question about this article? Send it to firstname.lastname@example.org and the response will be shared in MDT's blog online.
Isolation Class Input voltage 15 VDC 6 VDC 75VDC 150 VDC 300 VDC 12 VAC 30 VAC 60 VAC 125 VAC 250 VAC Basic Clearance 0.4mm 0.5mm 0.7mm 1.0mm 1.6mm Creepage 0.8mm 1.0mm 1.3mm 2.0mm 3.0mm Supplementary Clearance 0.8mm 1.0mm 1.2mm 1.6mm 2.5mm Creepage 1.7mm 2.0mm 2.3mm 3.0mm 4.0mm Reinforced Clearance 1.6mm 2.0mm 2.4mm 3.2mm 5.0mm Creepage 3.4mm 4.0mm 4.6mm 6.0mm 8.0mm Table 1: This table shows the defined air gap and creepage distances depending on the input voltage. The values for reinforced isolation are approximately three times greater than for basic isolation.
Jordi Torrebadell is the Director of Engineering - Americas at RECOM Power Inc. He is responsible for engineering support and development of RECOM products in North and South America. He can he reached at 718-855-9710 or email@example.com.
|Printer friendly Cite/link Email Feedback|
|Title Annotation:||Emphasis On Medical Electronics|
|Publication:||Medical Design Technology|
|Date:||Jan 1, 2012|
|Previous Article:||Challenging channels: successfully designing multi-lumen tubing.|
|Next Article:||Three ways to bond tubing to standard connectors.|