European concerted action on anticoagulation. Use of plasma samples to derive international sensitivity index for whole-blood prothrombin time monitors.Monitors for point-of-care testing (POCT POCT Point of care testing, see there )  of prothrombin time Prothrombin Time Definition
The prothrombin time test belongs to a group of blood tests that assess the clotting ability of blood. The test is also known as the pro time or PT test. (PT) in whole blood are being used on an increasing scale for near-patient and home PT testing. For safe oral anticoagulation, the International Normalized Ratio International Normalized Ratio Hematology A method of reporting prothrombin time–PT results for Pts receiving oral anticoagulant therapy; the INR is defined by the formula, PTPatient/PTMNPT (INR INR
In currencies, this is the abbreviation for the Indian Rupee.
The currency market, also known as the Foreign Exchange market, is the largest financial market in the world, with a daily average volume of over US $1 trillion. ) for all monitors must conform with the WHO PT standardization scheme (1, 2). This demands the calibration of monitors in terms of the International Sensitivity Index (ISI ISI International Sensitivity Index, see there ). A method for this has been described by Tripodi and coworkers (3, 4), but it is not feasible for routine work. They adapted the conventional WHO procedure for use with two different types of POCT monitors. Their method requires parallel testing on the monitors of noncitrated whole blood from 60 warfarin-treated patients and 20 healthy controls and of citrated plasmas from the same blood samples with the conventional manual PT technique. In the present study, whole blood was tested on three POCT whole-blood monitoring systems: CoaguChek Mini, CoaguChek-S, and Thrombolytic thrombolytic /throm·bo·lyt·ic/ (throm?bo-lit´ik) dissolving or splitting up a thrombus, or an agent that so acts.
1. dissolving or splitting up a thrombus.
2. an agent that dissolves or splits up a thrombus. Assessment System prothrombin time noncitrated (TAS TAS
1. telephone answering system
2. true airspeed PT-NC). In this report, a system is defined as the POCT monitor and a single lot of a specific type of test strips/ cards.
We also tested citrated plasmas recalcified with a calcium chloride calcium chloride, CaCl2, chemical compound that is crystalline, lumpy, or flaky, is usually white, and is very soluble in water. The anhydrous compound is hygroscopic; it rapidly absorbs water and is used to dry gases by passing them through it. concentration that we found suitable for PT testing of plasmas on the POCT systems (L. Poller, manuscript in preparation). The aim of the present study was to determine whether the ISIs of these monitors could be derived reliably with citrated plasmas. If successful, ISI calibration of whole-blood POCT monitors could then be simplified.
To avoid invidious in·vid·i·ous
1. Tending to rouse ill will, animosity, or resentment: invidious accusations.
2. comparison with the several different types of POCT monitors on the market that have not been subjected to the same analysis, results with the three POCT systems have been coded.
Materials and Methods
The CoaguChek and TAS systems, consisting of the monitors and specific lots of test strips/cards, were provided for the study by Roche Diagnostics (Mannheim, Germany) and Bayer (Leverkusen, Germany), respectively.
CoaguChek. The CoaguChek system consisted of the meter, test strips incorporating thromboplastin thromboplastin: see blood clotting. , and a code chip containing calibration information relevant to the specific lot of test strips, such as mean normal prothrombin time (MNPT MNPT Male National Pipe Thread ), ISI, lot number, and expiration date Expiration Date
The day on which an options or futures contract is no longer valid and, therefore, ceases to exist.
The expiration date for all listed stock options in the U.S. . For this system, the test strip, which contains iron oxide The material used to coat the surfaces of magnetic tapes and lower-capacity disks. particles and thromboplastin, is inserted into the monitor and warmed to 37[degrees]C. An unmeasured drop of capillary blood (noncitrated) is applied to the strip. The blood is then transported to the reaction area by capillary forces, and the coagulation process is triggered by contact with the thromboplastin. There are two magnets below the test strip, a permanent magnet and an electromagnet electromagnet, device in which magnetism is produced by an electric current. Any electric current produces a magnetic field, but the field near an ordinary straight conductor is rarely strong enough to be of practical use. . The former causes the iron particles on the test strip to align horizontally, and the latter forces these particles into the vertical plane within a set frequency, giving rise to a regular pulsation pulsation /pul·sa·tion/ (pul-sa´shun) a throb, or rhythmic beat, as of the heart.
1. The act of pulsating.
2. A single beat, throb, or vibration. pattern. A photodetector A device that senses light. It uses the principle of photoconductivity, which is exhibited in certain materials that change their electrical conductivity when exposed to light. See photoelectric, photocell and photodiode. above the test strip registers the changes in reflected light caused by this pulsation pattern. As soon as a clot begins to form, movement of the iron particles slows until it finally stops. This produces a decrease in reflection, which is interpreted as the onset of coagulation. An algorithm programmed into the specific code chip converts this result into PT and INR.
Two CoaguChek meters and single lots of two different types of test strips (CoaguChek-S test strip lot no. 828594 and CoaguChek Mini test strip lot no. 079) were provided to each of three centers. CoaguChek-S and Mini test strips contain human recombinant and rabbit plain thromboplastin, respectively.
TAS. The TAS has recently been redesignated the RapidPointCoag, but in this study the established name of TAS has been retained. The TAS is designed primarily for near-patient testing, but it can be used for home PT testing. The TAS works on essentially the same principles as the CoaguChek. The test carriers are referred to as "test cards". A magnetic strip on the back of each test card contains all relevant calibration information, type of test, lot number, expiration date, and other test parameters, such as MNPT and ISI, specific to the individual lot of cards. This information is entered into the TAS by passing the test card through the magnetic card reader at the beginning of each test.
The test cards (lot no. 305099801) used in this study are designated PT-NC (prothrombin time noncitrated) and contain human placental placental
pertaining to or emanating from placenta.
the placental separation of maternal and fetal blood which varies in its structure and permeability between the species. plain thromboplastin.
Until the 1990s, thromboplastins with a high ISI (>2.0) were widely used in North America, and many laboratories were accustomed to the resulting short PT. PTs displayed on some POCT monitors were therefore in terms of a typical North American North American
named after North America.
North American blastomycosis
see North American blastomycosis.
North American cattle tick
see boophilusannulatus. thromboplastin, which are not real measurements in seconds.
The CoaguChek Mini displays PTs that are considered representative of the result, in seconds, that would have been obtained with a traditional high-ISI thromboplastin. To provide results in real PT in this study, a master code chip containing a correction formula was provided by Roche Diagnostics for the CoaguChek Mini system rather than the conventional code chip.
The TAS PT-NC also does not give real PT results. During the development of the PT-NC test cards, the objective of the manufacturer was to have results as close as possible to laboratory results. The PTs on the TAS PT-NC system are therefore expressed as an assumed laboratory PT result with the actual TAS PT being longer than the displayed result on the monitor. For the TAS PT-NC, a specific correction formula was supplied by the manufacturer for this study for calculation of real PTs from those reported on the monitor display screen.
THREE-CENTER WHO ISI CALIBRATION
A WHO ISI calibration as modified for POCT monitors by Tripodi et al. (3) was conducted at three centers: Leiden, Manchester, and Milan. Each center collected blood samples from 20 healthy blood donors and 60 long-term patients stabilized on oral anticoagulants within the therapeutic interval (INR, 1.5-4.5). Blood samples were collected over a 5- to 10-day period at each center. All volunteers gave informed consent.
COLLECTION AND TESTING OF NONCITRATED WHOLE BLOOD ON THE MONITORS
Noncitrated whole blood was collected by venipuncture venipuncture /veni·punc·ture/ (ven?i-pungk´chur) surgical puncture of a vein.
ve·ni·punc·ture or ve·ne·punc·ture
n. using a needle and plastic syringe. The blood was applied with minimum delay to the test strips/cards in the monitors, and the PTs were recorded from the monitor display screens. Single tests were performed on all whole-blood samples.
BLOOD COLLECTION FOR CITRATED PLASMA SAMPLES
An additional sample of venous blood venous blood
n. Abbr. v
Blood that has passed through the capillaries of various tissues other than the lungs, is found in the veins, in the right chambers of the heart, and in pulmonary arteries, and is usually dark red as a result of a from the same donor was collected at the same venipuncture, either with a Vacutainer system containing 0.105 mol/L citrate citrate /cit·rate/ (sit´rat) a salt of citric acid.
citrate phosphate dextrose (CPD) anticoagulant citrate phosphate dextrose solution. or with a Monovette system containing 0.106 mol/L citrate (the difference in citrate concentration between these systems is negligible). Citrated plasma was obtained by centrifugation Centrifugation
A mechanical method of separating immiscible liquids or solids from liquids by the application of centrifugal force. This force can be very great, and separations which proceed slowly by gravity can be speeded up enormously in centrifugal and stored at room temperature before testing. All tests were completed within 6 h of venipuncture.
TESTING OF CITRATED PLASMA ON THE MONITORS
Citrated plasma (0.1 mL) was transferred to a plastic tube. Calcium chloride (0.1 mL; 16.3 mmol/L) was added and mixed to give a clotting time comparable to the whole-blood clotting time and the traditional manual PT method.
Recalcified plasma was applied within 15 s  to the test strip/cards, and the PTs were recorded from the monitor display screens. Single tests were performed on the monitoring systems with all plasma samples. The 16.3 mmol/L calcium chloride solution was prepared individually at each of the three centers.
Manual PTs for the citrated plasmas were obtained with the European Concerted Action on Anticoagulation (ECAA ECAA European Common Aviation Area
ECAA Egyptian Civil Aviation Authority
ECAA Esophageal Cancer Awareness Association
ECAA Echelon Corps And Above ) rabbit reference plain thromboplastin ([ISI.sub.ref] = 1.67) according to the recommended procedure (5).
Single tests were performed with the patients' plasmas and duplicate tests with the control plasmas.
Calibration data. The ISIs for whole blood and citrated plasma with the three monitoring systems were derived according to the conventional WHO orthogonal regression procedure (1, 2). The logs of individual PTs for healthy blood donors and coumarin-treated patients, respectively, obtained with the ECAA rabbit reference reagent were plotted as recommended on the vertical axis against the log of the PT obtained from the monitor on the horizontal axis (1, 2). The slope of the calibration line was calculated using orthogonal regression analysis In statistics, a mathematical method of modeling the relationships among three or more variables. It is used to predict the value of one variable given the values of the others. For example, a model might estimate sales based on age and gender. . The precision of the calibration slope (b) was determined by its CV: CV(b) = 100 = SE(b)/b, where SE (b) is the standard error of b. The assumption that the mean log PT for the healthy blood donors lies on the calibration line derived from the PTs for patient samples only was assessed by the displacement test of Tomenson (6). This determines the significance of the vertical distance (displacement) of the mean of the log PT for the healthy blood donors from the patient-only line. The estimated slopes of the calibration lines based on the PTs for both healthy blood donors and patients and the corresponding 95% confidence intervals are given to allow a comparison between whole-blood and citrated plasma calibrations for each of the three monitoring systems. The ISI was derived as: ISI = b x [ISI.sub.ref], where [ISI.sub.ref] = 1.67 is the ISI of the ECAA rabbit reference thromboplastin.
MNPT. The MNPTs (geometric mean (mathematics) geometric mean - The Nth root of the product of N numbers.
If each number in a list of numbers was replaced with their geometric mean, then multiplying them all together would still give the same result. of the PTs obtained for the 20 samples from healthy blood donors) for whole blood and citrated plasma were compared using paired t-tests on the log PT obtained for the healthy blood donors, for the three centers, and for the three monitoring systems.
Assessment of INR. The prothrombin prothrombin
Carbohydrate-protein compound in plasma essential to coagulation. In response to bleeding, a complex series of clotting-factor interactions leads to its conversion by thromboplastin to thrombin, which transforms fibrinogen in plasma into fibrin. ratio (PR), which corresponds theoretically to an INR of 3.0 (this being a good representative degree of anticoagulation) with the mean whole-blood ISI of the three centers, was calculated. The INRs derived from this PR and the mean citrated plasma ISI were then obtained, and the absolute percentage of deviation from the whole-blood INR of 3.0 was determined. As in previous reports (7-9), the absolute percentage of INR deviation was deemed clinically relevant if >10%.
[FIGURE 1 OMITTED]
Results for the three monitoring systems are coded as systems A, B, and C. The ISI values and the calibration statistics for whole-blood samples and plasmas on the three monitoring systems at each of the three centers are given in Table 1. The mean results for these three centers are given in Table 2.
The CV of the calibration line slopes was 1.9-3.9% excluding system C results at Milan, which used a different lot number of test cards/strips. For systems A and C at all three centers and system B at two centers, the CV of the slope was smaller for plasmas than for whole-blood samples. For system A at all three centers, there was a small difference between the CVs of the slopes for the whole-blood samples and plasmas. There were larger differences between the whole-blood and citrated plasma CVs at one of the three centers with system B and at all three centers with system C.
There was evidence (P <0.01) that the mean log PT for healthy individuals did not lie on the patient-only calibration line (i.e., significant displacement) in 2 of the 18 calibrations. This is greater than the number of significant displacements that would be expected by chance in 18 tests at the 1% level. The two calibrations giving significant displacement were with plasmas on systems A and B (see Table 1). Fig. 1 shows both the lines for patients and healthy blood donors combined and the patient-only line for these calibrations.
[FIGURE 2 OMITTED]
The clinical significance of the displacement was assessed according to WHO guidelines (2). The PR corresponding to a theoretic Tomenson-corrected (6) INR of 2.0 was calculated. The INR determined with this PR and the ISI derived from the slope of the line for samples from both patients and healthy individuals was compared with the INR of 2.0 (see Table 1). This was repeated for a theoretic INR of 4.5. The absolute percentage of INR deviation exceeded the WHO 10% limit (2) for system B only.
Again there was little difference in mean ISI (see Table 2) for whole-blood and plasma samples at the three centers on systems A and B. However, for system C, the difference between the mean ISIs for whole blood and citrated plasma was much greater. Fig. 2, which gives the calibration slopes and the 95% confidence intervals, allows a comparison of slopes for whole blood and citrated plasma. Although there were between-center differences in the slopes for both systems A and B, there was little within-center difference between the slopes obtained for whole blood and plasma at the individual centers. With system C, the within-center difference between the calibration slopes for whole blood and plasma was much greater at two of the three centers.
The theoretic citrated plasma INR derived from the mean plasma ISI and the PR corresponding to an INR of 3.0 obtained from the mean whole-blood ISI are shown in Table 2. The absolute percentage of deviation of the plasma INR from the whole-blood INR of 3.0 was considerably less than the 10% limit for systems A and B. For system B, there was no difference between INRs because the mean ISIs for whole blood and citrated plasma were the same. With system C, the two centers with the same lot of test cards/strips gave ~12% lower INRs with plasma than with whole blood.
The MNPTs with the three monitoring systems based on both whole-blood and plasma samples are given in Table 3. At all three centers with systems A and B, MNPTs were significantly longer (P <0.0001) with plasmas than with whole blood. With system C, the pattern in the difference was not consistent at the three centers. At Leiden, the MNPT was significantly shorter (P <0.0001) with the plasmas than with the whole blood. At Manchester, the plasma MNPT on system C was shorter than the MNPT for whole blood, but the difference was not significant (P = 0.06). The plasma MNPT was significantly longer (P = 0.03) than the MNPT for whole blood at Milan.
This study demonstrates that provided an appropriate calcium chloride concentration is used to recalcify the citrated plasma, plasma PT results can be used for accurate ISI calibration of two of the three whole-blood POCT monitoring systems considered in this report. Results showed little difference between plasma and whole-blood ISIs. With the third system, the difference in ISI between plasma and whole-blood calibrations produced a clinically relevant difference in INR. A theoretic PR of 2.6 gave a 12% INR difference between whole blood and citrated plasma (3.00 and 2.63, respectively). This would need to be investigated further to determine whether this problem can be resolved.
The precision of the resulting calibrations using the WHO procedure modified by Tripodi et al. (3) based on 20 healthy controls and 60 long-term coumarin-treated patients was satisfactory. The CV of the calibration slope was less than the recommended upper limit of 3% set for conventional manual PT testing (2) at two centers on two monitoring systems with both whole-blood and citrated plasma samples. At the third center, the slope CVs were close to 3%. With the third monitoring system, slope CVs were <3% at one center and <5% at all three centers. There was no suggestion, however, that substitution of plasma samples made the precision of the calibration poorer with the PT monitors; indeed, the mean CV of the slope was greater with whole blood for all three monitoring systems.
In the present study, a secondary reference thromboplastin (ECAA rabbit reference plain thromboplastin) was used in the ISI calibrations. Although WHO guidelines recommend like-species calibrations (2), only one of the three POCT systems was calibrated with like-species thromboplastin. For this reason and because of the important differences between results at the three centers, a larger study is required with relevant WHO International Reference Preparations for accurate ISI determinations. This is currently being undertaken in an ECAA multicenter calibration of the CoaguChek Mini and TAS systems against the human and rabbit thromboplastin International Reference Preparations.
The members of the ECAA Steering Group are as follows: J. Jespersen (Chairperson), L. Poller (Co-ordinator), A.M.H.P. van den Besselaar, F.J.M. van der Meer Van der Meer is a Dutch surname that simply means the phrase 'from the lake' in English. Many years ago, descendants would have lived from a lake in the Netherlands which is how the name first originated. , C. Shiach, A. Tripodi, and F.E. Preston (Consultant). The study was supported by EC Grant SMT (1) (Surface Mount Technology) See surface mount.
(2) (Station ManagemenT) An FDDI network management protocol that provides direct management. Only one node requires the software.
SMT - Station Management 4-CT98-2269 and an additional grant from the Manchester Thrombosis Research Foundation. We express our gratitude to Roche Diagnostics and Bayer for the loan of the CoaguChek and TAS instruments, respectively, and test strips/cards.
(1.) WHO Expert Committee on Biological Standardization. 33rd report. World Health Organ Tech Rep Ser 1983;687:81-105.
(2.) WHO Expert Committee on Biological Standardization. Guidelines for thromboplastins and plasma used to control oral anticoagulation therapy. Annex 3. World Health Organ Tech Rep Ser 1999; 889:64-93.
(3.) Tripodi A, Arbini AA, Chantarangkul V, Bettega D, Mannucci PM. Are capillary whole blood coagulation monitors suitable for the control of oral anticoagulant treatment by the International Normalized Ratio? Thromb Haemost 1993;70:921-4.
(4.) Tripodi A, Chantarangkul V, Clerici M, Negri B, Mannucci PM. Determination of the ISI of a new near-patient testing device to monitor oral anticoagulant therapy. Thromb Haemost 1997;78:855-8.
(5.) Poller L, Barrowcliffe TW, van den Besselaar AMHP AMHP AmeriHealth Mercy Health Plan , Jespersen J, Houghton D. European Concerted Action on Anticoagulation (ECAA)-the multicentre calibration of rabbit and human ECAA reference thromboplastins. Steering Committee. Thromb Haemost 1996;76:977-82.
(6.) Tomenson JA. A statistician's independent evaluation. In: van den Besselaar AMHP, Lewis SM, Gralnick HR, eds. Thromboplastin calibration and oral anticoagulant control. Boston: Martinus Nijhoff, 1984:87-108.
(7.) Poller L, Barrowcliffe TW, van den Besselaar AMHP, Jespersen J, Tripodi A, Houghton D, for the ECAA. Minimum lyophilized plasma requirement for ISI calibration. Am J Clin Pathol 1998;109:196-204.
(8.) Poller L, van den Besselaar AMHP, Jespersen J, Tripodi A, Houghton D, for the ECAA. The effect of sample size on fresh plasma thromboplastin ISI determination. Br J Haematol 1999; 105:655-63.
(9.) Poller L. Screening INR deviation of local prothrombin time systems. J Clin Pathol 1998;51:356-9.
 Nonstandard non·stan·dard
1. Varying from or not adhering to the standard: nonstandard lengths of board.
2. abbreviations: POCT, point-of-care testing; PT, prothrombin time; INR, International Normalized Ratio; ISI, International Sensitivity Index; TAS, Thrombolytic Assessment System; PT-NC, prothrombin time, noncitrated; MNPT, mean normal PT; ECAA, European Concerted Action on Anticoagulation; and PR, prothrombin ratio.
 To determine the maximum time allowable between recalcification and testing of plasma samples on the POCT monitoring systems, recalcified plasma was added to two of the three POCT systems after 5, 30, 60, and 90 s with the order of testing on the two instruments being reversed on subsequent samples. Five replicate tests were performed for each time interval. Results obtained on the CoaguChek system showed no change over the time intervals, but results obtained on the TAS system suggested some shortening of PT after 30 s. Detailed study on the TAS performed at intervals of 5 s for up to 30 s indicated a progressive shortening of PT after 15 s. It was therefore decided to perform all tests with recalcified plasmas on all three POCT systems within 15 s.
LEON POLLER,  * MICHELLE MICHELLE Mid-Infrared Echelle Spectrograph KEOWN,  NIKHIL CHAUHAN,  ANTON M.H.P. VAN DEN BESSELAAR,  JOYCE MEEUWISSE-BRAUN,  ARMANDO TRIPODI,  MARIAGRAZIA CLERICI,  AND JORGEN JESPERSEN 
 European Concerted Action on Anticoagulation (ECAA) Central Facility, School of Biological Sciences, The University of Manchester The University of Manchester is a university located in Manchester, England. With over 40,000 students studying 500 academic programmes, more than 10,000 staff and an annual income of nearly £600 million it is the largest single-site University in the United Kingdom and receives , Manchester M13 9PT, United Kingdom.
 Haemostasis hemostasis, haemostasis
the stoppage of bleeding or cessation of the circulation of the blood; stagnation of the blood in a part of the body. Also hemostasia, haemostasia.
See also: Blood and Blood Vessels
Noun 1. and Thrombosis Research Center, Leiden University Medical Center The Leiden University Medical Center (Dutch: Leids Universitair Medisch Centrum) or LUMC, is the university hospital affiliated with Leiden University, of which it forms the medical faculty. , 2333ZA Leiden, The Netherlands.
 A Bianchi Bonomi, Haemophilia and Thrombosis Centre, IRCCS IRCCS Istituto Di Ricovero e Cura a Carattere Scientifico (Italian Research Hospital) Maggiore Hospital, University of Milan The university is a member of the League of European Research Universities.
Throughout Milan, the University is normally known as Statale to avoid confusion with other academic institutions in the city. , 20122 Milan, Italy.
 Department for Thrombosis Research, University of Southern Denmark As a national institution the University of Southern Denmark (SDU) comprises five faculties – Humanities, Science, Engineering, Social Sciences and Health Sciences totaling 32 departments, 11 research centers and a university library. , and Department of Clinical Chemistry, Ribe County Hospital, DK-6700 Esbjeig, Denmark.
* Author for correspondence. Fax 44-161-275-5316; e-mail email@example.com. Received July 23, 2001; accepted November 1, 2001.
Table 1. ISI and calibration statistics for individual centers. (a) Center Test Slope CV(b), ISI (b) % System A Leiden Whole blood 0.97 2.1 1.62 Plasma 0.93 1.9 1.56 Manchester Whole blood 0.90 3.2 1.50 Plasma 0.95 3.2 1.58 Milan Whole blood 0.85 2.9 1.42 Plasma 0.79 2.9 1.32 (b) (4%, 8%) System B Leiden Whole blood 0.72 2.5 1.20 Plasma 0.69 1.9 1.15 (b) (28%, 37%) Manchester Whole blood 0.65 2.8 1.08 Plasma 0.65 2.8 1.08 Milan Whole blood 0.58 2.9 0.97 Plasma 0.60 2.8 1.00 System C (c) Leiden Whole blood 0.74 2.8 1.24 Plasma 0.63 2.1 1.06 Manchester Whole blood 0.63 3.9 1.06 Plasma 0.58 3.2 0.96 Milan3 Whole blood 0.72 4.7 1.20 Plasma 0.61 3.2 1.01 (a) Estimated slope (b), CV (%) of b [CV(b)], and ISI for each monitor test system with noncitrated whole-blood and plasma samples. (b) Significant displacement at the 1% level using the Tomenson (6) test (the absolute percentages of INR deviation from theoretic INRs of 2.0 and 4.5, respectively, are given in parentheses). (c) Different lot of test cards/strips used with system C monitor at Milan. Table 2. Mean calibration statistics. (a) Monitoring Test Mean Mean Mean INR %diff system slope CV(b), % ISI A Whole blood 0.91 2.8 1.51 3.00 -1.7 Plasma 0.89 2.7 1.49 2.95 B Whole blood 0.65 2.7 1.08 3.00 0.0 Plasma 0.65 2.6 1.08 3.00 C (b) Whole blood 0.69 3.4 1.15 3.00 -12.3 Plasma 0.61 2.7 1.01 2.63 (a) Mean slopes, mean CV of slopes, and mean ISI obtained for each monitor test system with samples of noncitrated whole blood and citrated plasma. Theoretic plasma INR determined from the mean plasma ISI and the PR corresponding to a whole-blood INR of 3.0 are given with the percentage of deviation (%diff) of plasma INR from the whole-blood INR of 3.0. (b) Mean of results for two centers only. Table 3. MNPT and SD (in parentheses) for 20 healthy blood donors. (a) Mean (SD) MNPT Monitoring Test Leiden Manchester system System A Whole blood 17.7 (1.34) 17.3 (1.05) Plasma 19.3 (1.68) 19.9 (1.11) System B Whole blood 12.5 (0.57) 12.1 (0.57) Plasma 13.6 (0.72) 13.1 (0.97) System C (b) Whole blood 15.3 (1.10) 14.3 (1.15) (c) Plasma 13.6 (1.10) 13.7 (1.07) ECAA rabbit reference Plasma 16.6 (1.25) 17.5 (0.73) thromboplastin Mean (SD) MNPT Monitoring Test Milan Overall system MNPT (SD) System A Whole blood 18.5 (1.02) 17.8 (1.23) Plasma 20.9 (1.53) 20.0 (1.58) System B Whole blood 12.2 (0.69) 12.3 (0.62) Plasma 13.6 (0.76) 13.5 (0.83) System C (b) Whole blood 12.9 (0.82) 14.9 (1.21) Plasma 13.4 (0.68) 13.7 (1.07) ECAA rabbit reference Plasma 18.8 (1.34) 17.7 (1.45) thromboplastin (a) Samples were collected locally and tested as both whole-blood and plasma samples on each monitoring test system. Plasma samples were also tested with ECAA rabbit reference thromboplastin using the manual PT method. The overall MNPT is the geometric mean of PT results for all centers. (b) Overall MNPT (SD) for system C excludes Milan results because a different lot of test strips was used. (c) MNPT (SD) calculated from PT results for 19 healthy blood donors.