Development of a tunable LED-based colorimetric source.A novel, spectrally tunable light-source utilizing light emitting diodes (LEDs) for radiometric, photometric pho·tom·e·try
Measurement of the properties of light, especially luminous intensity.
photo·met , and colorimetric col·or·im·e·ter
1. Any of various instruments used to determine or specify colors, as by comparison with spectroscopic or visual standards.
2. applications is described. The tunable source can simulate standard sources and can be used as a transfer source to propagate photometric and colorimetric scales from calibrated cal·i·brate
tr.v. cal·i·brat·ed, cal·i·brat·ing, cal·i·brates
1. To check, adjust, or determine by comparison with a standard (the graduations of a quantitative measuring instrument): reference instruments to test artifacts artifacts
see specimen artifacts. with minimal increase in uncertainty. In this prototype source, 40 LEDs with 10 different spectral distributions were mounted onto an integrating sphere. A voltage-to-current control circuit was designed and implemented, enabling independent control of the current sent to each set of four LEDs. The LEDs have been characterized for stability and dependence on drive current. The prototype source demonstrates the feasibility of development of a spectrally tunable LED source using LEDs with up to 40 different spectral distributions, Simulations demonstrate that such a source would be able to approximate standard light-source distributions over the visible spectral range--fro m 380 nm to 780 nm--with deviations on the order of 2 %. The tunable LED source can also simulate spectral distributions of special sources such as discharge lamps and display monitors. With this tunable source, a test instrument can be rapidly calibrated against a variety of different source distributions tailored to the anticipated uses of the artifact. Target uncertainties for the calibration of test artifacts are less than 2 % in luminance The amount of brightness, measured in lumens, that is given off by a pixel or area on a screen. For example, dark red and bright red would have the same chrominance, but a different luminance. and 0.002 in chromaticity for any source distribution.
Key words: colorimetry colorimetry
Measurement of the intensity of electromagnetic radiation in the visible spectrum transmitted through a solution or transparent solid. It is used to identify and determine the concentrations of substances that absorb light of a specific wavelength or colour ; integrating sphere; light emitting diode; photometry photometry (fōtŏm`ətrē), branch of physics dealing with the measurement of the intensity of a source of light, such as an electric lamp, and with the intensity of light such a source may cast on a surface area. ; radiometry Radiometry
A branch of science that deals with the measurement or detection of radiant electromagnetic energy. Radiometry is divided according to regions of the spectrum in which the same experimental techniques can be used. .
There has been a great deal of interest in the development of a stable, spectrally tunable, radiometric source for a variety of photometric, colorimetric and radiometric programs. For example, commercial integrating sphere sources typically utilize broadband sources such as incandescent lamps or Xe-arc sources with fixed spectral distributions. Colorimeters can be calibrated against these sources with minimal errors, on the order of 0.001 in chromaticity (x,y) and 1 % in luminance (Y). However, errors in measurements of artifacts with different spectral distributions--e.g., colored sources such as displays--using these colorimeters can be an order of magnitude A change in quantity or volume as measured by the decimal point. For example, from tens to hundreds is one order of magnitude. Tens to thousands is two orders of magnitude; tens to millions is three orders of magnitude, etc. larger. Ideally, these colorimeters should be calibrated using light sources with spectral distributions similar to the distributions of test artifacts measured by the colorimeters. Development of a spectrally tunable source would enable rapid calibration of these colorimeters when measuring different spectral distributions. Additional applications inc lude lude
A pill or tablet that contains the drug methaqualone. evaluation of material properties under differing illumination conditions, characterization of reflective materials and displays, and calibrations of instruments that measure nonstandard non·stan·dard
1. Varying from or not adhering to the standard: nonstandard lengths of board.
2. spectral distributions. A spectrally tunable source using a conventional lamp and monochromator A monochromator is an optical device that transmits a mechanically selectable narrow band of wavelengths of light or other radiation chosen from a wider range of wavelengths available at the input. with a multi-element liquid crystal filter is currently under development (1).
Recent advances in materials and manufacture have resulted in the commercial availability of light emitting diodes (LEDs) with narrow spectral distributions, high power, and dominant wavelengths spanning the entire visible spectrum (2). The radiometric and photometric properties of a variety of LEDs have been measured (3), and they are being further evaluated as possible standard sources (4,5). By equipping an integrating sphere with a number of LEDs having different spectral distributions and varying the radiometric output of the various LEDs, the sphere radiance can be tuned from the blue to the red, as well as tuned to approximate CIE-defined standard illuminant A standard illuminant is a profile or spectrum of visible light which is published in order to allow images or colours recorded under different lighting to be compared. CIE illuminants
The International Commission on Illumination (usually abbreviated CIE distributions (e.g., Illuminant il·lu·mi·nant
Something that gives off light.
[Latin illmin A or D65), enabling calibrations of colorimeters and spectro-radiometers against different spectral distributions. Detector-based photometric and tristimulus colorimetric scales have been realized on filtered trap-detectors where spectral corrections were introduced to perform high accuracy illuminance illuminance: see photometry.
A term expressing the density of luminous flux incident on a surface. This word has been proposed by the Colorimetry Committee of the Optical Society of America to replace the term illumination. and color mea surements of sources with widely varying spectral distributions (6). The high accuracy scales can be transferred to working standard and test meters utilizing the tunable LED source and adjusting its spectral distribution to approximate that of a specific test source. In this work, we present simulations of an idealized i·de·al·ize
v. i·de·al·ized, i·de·al·iz·ing, i·de·al·iz·es
1. To regard as ideal.
2. To make or envision as ideal.
1. LED-based source and its ability to reproduce standard illuminant distributions along with the design and characterization of an LED-based integrating sphere source.
Fig. 1, we show the results of simulations of an idealized LED source with 20 and 40 LEDs having Gaussian spectral distributions with 20 nm widths spanning the colorimetric spectral range from 380 nm to 780 nm. The simulations demonstrate that incorporation of 40 LEDs with different spectral distributions into a single unified source could enable sufficient reproduction of standard illuminant sources and possibly common display phosphor A rare earth material used to coat the inside face of a CRT. When struck by an electron beam, the phosphor emits a visible light for a few milliseconds. In color displays, red, green and blue phosphor dots are grouped as a cluster. See screen burn. distributions.
We recently developed a prototype source to evaluate the potential development of LED-based tunable integrating sphere sources (ISS ISS
See Institutional Shareholder Services (ISS). ) for photometric, colorimetric and radiometric applications. The prototype ISS described in this work is equipped with a total of 40 LEDs, 4 LEDs each with 10 different spectral distributions. We describe the design and characterization of the electronic circuit as well as the aging characteristics of typical red, green and blue LEDs used in the source.
3. ISS Design and Layout
A schematic diagram of the LED source is shown in Fig. 2. A 10-channel electronic control box determines the current sent to each set of 4 LEDs. Twenty LEDs are included in each of two source modules that mount on either side of a commercial integrating sphere. Two LEDs in each module are connected in series and controlled with one channel in the electronic control box. The source is designed to be stable, but, because of its tunability, it is not developed as a standard calibrated source with long term reproducibility. Instead, two monitor detectors, a single channel photopic detector and a diode array spectroradiometer, are used to determine the real-time photometric or spectral radiometric output of the ISS. A computer logs the photometric or radiometric output of the monitor detector and can turn the LEDs on or off. In the future, the current to each set of LEDs will be remotely set by a computer through the General Purpose Interface Bus General Purpose Interface Bus - IEEE 488 (GPIB (General Purpose Interface Bus) An IEEE 488 standard parallel interface used for attaching sensors and programmable instruments to a computer. Using a 24-pin connector, up to 15 devices can be daisy chained together. HP's version is the HPIB. ).
3.1 Electronic Control Box
A simplified schematic diagram of the electronic control-box is shown in Fig. 3. Ten voltage-to-current (V-I) converter circuits provide a constant current to 10 sets of 4 LEDs. The current of each LED drive circuit can be switched on and off in either manual or remote control mode. Also, the input voltages, [V.sub.1] to [V.sub.10], of the V-I converters can be controlled in both operational modes (manual or remote). Two power supplies are used. A [+ or -] 14.5 V reference voltage source A voltage source is any device or system that produces an electromotive force between its terminals OR derives a secondary voltage from a primary source of the electromotive force. feeds the power stages of the V-I converters, while the other circuits are operated from a [+ or -] 15 V power supply.
The circuit diagram A circuit diagram (also known as an electrical diagram, elementary diagram, or electronic schematic) is a simplified conventional pictorial representation of an electrical circuit. of a V-I converter is shown in detail in Fig. 4. The input voltage [V.sub.1] is attenuated Attenuated
Alive but weakened; an attenuated microorganism can no longer produce disease.
Mentioned in: Tuberculin Skin Test
having undergone a process of attenuation. from a Zener diode Zener diode: see diode.
A two-terminal semiconductor junction device with a very sharp voltage breakdown as reverse bias is applied. The device is used to provide a voltage reference. It is named after C. by a 10-turn potentiometer equipped with a reading dial. Alternatively, the output voltage of a D/A converter (Digital/Analog converter) A device that converts digital code into analog signals. The most common applications are for generating sound and video. The D/A converter may be contained on a single chip or can be one circuit within a chip. Contrast with A/D converter. can be used in remote control mode. The input voltage [V.sub.1] controls the output voltage [V'.sub.1] through a two stage amplifier. The first stage is an inverting voltage amplifier. The second stage is a voltage follower for the load resistor ([R.sub.L]) that works as a current source for the serially connected LEDs. Both stages utilize analog control loops (using negative feedback from the output back to the input) to obtain high signal gain accuracy and stability. The highly stable supply-voltage of the output current source is 0.5 V lower than the + 15 V power supply of the second-stage control circuit. Consequently, [V'.sub.1] does not have to be higher than 14.5 V. The 0.5 V difference allows [V'.sub.1] to follow the output voltage of the first stage with a high accuracy ev en for very small LED current selections. The current of the LEDs can be easily switched on and off using reed relays. The TTL (1) (Time To Live) A parameter in a network packet that sets a time limit to its validity. In order to prevent an IP packet from propagating endlessly through the network, the value in the TTL field is reduced by each router. input of the reed relays can be controlled in both manual and remote modes. The LED current is determined by the load resistor [R.sub.L] and the voltage difference between its terminals:
I = 14.5 V - [V'.sub.1]/[R.sub.L]. (1)
The LED current will be stable if the (14.5 V - [V'.sub.1]) difference and the load resistance remain constant. A high wattage wattage
the output or consumption of an electric device expressed in watts. resistor with a small temperature coefficient was selected for [R.sub.L].
3.2 Optical Head Design
The LED source module was designed to fit onto existing ports in a commercial integrating sphere. The source module was designed to enable rapid change of a particular set of LEDs within a module, or for the entire module to be easily changed. Two sets of source modules were fabricated. One set of source modules was equipped with LEDs having spectral distributions ranging from the blue to the red. The spectral distribution and chromaticity coordinates of a number of LEDs have been measured (2). Their chromaticity coordinates are shown in Fig. 5. From these LEDs, 10 sets of 4 LEDs with chromaticity coordinates varying from the blue to the red were mounted in the source module. Their chromaticity coordinates are also shown in Fig. 5.
3.3 Monitor Detectors
The integrating sphere source has one detector port. There are two interchangeable monitor detectors, a single element silicon detector with a photopic filter for monitoring the photometric output of the source and a commercial fiber-coupled diode-array spectroradiometer for continuously monitoring the spectral output of the source over the range from 350 nm to 860 nm.
4. Source Characterization
The electrical and optical properties of the LED source have been characterized. Each is discussed in detail below.
4.1 Electronic Characterization
The range of the control voltages [V.sub.1] and [V'.sub.1] for a load resistor of about 200 [ohm ohm (ōm) [for G. S. Ohm], unit of electrical resistance, defined as the resistance in a circuit in which a potential difference of one volt creates a current of one ampere; hence, 1 ohm equals 1 volt/ampere. ] is shown in Fig. 6. As the graph shows, the relation between the LED current and the control voltages is linear. The LED current can be regulated from 0 mA to 50 mA with a resolution of approximately 0.01 mA. Figure 7 shows the results of the stability test of the voltage-to-current converter. The output current to the selected LEDs was stable to better than 0.03 % of the set point (in this case 35 mA) over 14 h. Similar studies have been extended to 100 h with similar stability. No long term drifts in the current have been observed over a total operational time of 500 h.
4.2 Optical Characterization
The optical characteristics of representative red, green and blue LEDs used in the source are presented. The chromaticity coordinates of these LEDs are circled in Fig. 5. Each set of 4 LEDs was aged for 100 h, and their chromaticity and luminance monitored for the next 50 h of operation. For each set, the chromaticity was stable to within 0.001 in x and y, well within our target chromaticity uncertainty of 0.002. The measured luminance from the green and blue LEDs was stable to within 0.5 %, also well within our target uncertainty of 2 %, while the red luminance changed by 4 %. Further aging studies of the red LEDs are currently under way to establish the longer term stability of those LEDs. The source is stable to better than 0.2 % in all cases over the duration of several hours, the typical duration for a calibration.
The relationship between the luminance and the drive current of each set of LEDs was established, as shown in Fig. 8. The normalized relationship was very similar for the three sets of LEDs. The luminance of each set of LEDs approaches the luminance of a CRT (1) (C RunTime) See runtime library.
(2) (Cathode Ray Tube) A vacuum tube used as a display screen in a computer monitor or TV. The viewing end of the tube is coated with phosphors, which emit light when struck by electrons. display set to saturated red, green, and blue colors. With 50 mA supplied to all ten sets of LEDs, the source luminance was approximately 1000 cd/[m.sup.2] and its integrated radiance was approximately 500 [mu]W/c[m.sup.2]/sr. Incorporating newly developed higher intensity green and red LEDs, the source output can easily be doubled or tripled. The radiometric and photometric output of the source is comparable to that of a conventional lamp-illuminated integrating sphere source in the visible spectral region (7). Note that the LED source has negligible output in the infrared spectral region. In contrast, the output from a lamp-illuminated integrating sphere source with a color temperature of 3000 K peaks in the infrared at 1000 nm and has significant ra diant flux in the spectral range from 1000 nm to 2500 nm.
The chromaticity coordinates of the different LEDs were a strong function of drive current, as shown in Fig. 9. These changes arise from shifts in the spectral distribution of the LED with drive current. In developing the algorithm to approximate a specific source distribution, the spectral shift of each set of LEDs must be measured as a function of drive current and included in the model.
We have developed a prototype spectrally tunable source utilizing 40 LEDs with 10 different spectral distributions. A voltage-to-current control circuit was designed and implemented, enabling independent control of the current sent to each set of LEDs. The LEDs have been characterized for stability and dependence on drive current. The prototype source demonstrates the feasibility of development of a high-brightness, spectrally tunable LED source with independent control of up to 40 different sets of LEDs. Simulations demonstrate that such a source would be able to approximate the spectral distributions of standard illuminants over the visible range--from 380 nm to 780 nm--with errors on the order of 2 % in luminance and 0.002 in chromaticity. The tunable LED source will be useful in transferring photometric and colorimetric scales from reference instruments to test artifacts with a minimum increase in uncertainty for specific spectral distributions.
[FIGURE 1 OMITTED]
[FIGURE 2 OMITTED]
[FIGURE 3 OMITTED]
[FIGURE 4 OMITTED]
[FIGURE 5 OMITTED]
[FIGURE 6 OMITTED]
[FIGURE 7 OMITTED]
[FIGURE 8 OMITTED]
[FIGURE 9 OMITTED]
The authors would like to thank NIST (National Institute of Standards & Technology, Washington, DC, www.nist.gov) The standards-defining agency of the U.S. government, formerly the National Bureau of Standards. It is one of three agencies that fall under the Technology Administration (www.technology. colleagues Yoshi Ohno and C. Cameron Miller for useful discussions and the generous loan of the LEDs used in this work. C. Santana acknowledges support of the National Science Foundation through the Summer Undergraduate Research Foundation.
Accepted: June 19, 2002
Available online: http://www.nist.gov/jres
(1.) C. F. Wall, A. R. Hanson, and J. A. F. Taylor, Construction of a Programmable Light Source for Use as a Display Calibration Artefact See artifact. , Proc. SPIE SPIE International Society for Optical Engineering
SPIE Society of Photo-Optical Instrumentation Engineers
SPIE Source Path Isolation Engine
SPIE Special Purpose Insertion Extraction
SPIE Software Process Improvement Experimentation
SPIE Standard Protocols in Effect 4295, 259-266 (2001).
(2.) K. Muray, B. Kranicz, Y. Ohno, and J. Schanda, Comparison Measurements of LEDs: Spectral power distribution In color science, the power per unit area per unit wavelength of a radiant object. Mathematically, one may write:
where , Proc. CIE (Commission Internationale de l'Eclairage, International Commission on Illumination, Vienna, Austria, www.cie.co.at) An international organization that sets standards for all aspects of lighting and illumination, including colorimetry, photometry and the measurement of visible and 2nd Expert Symposium on LED Measurements, Gaithersburg, MD (2001) pp. 52-55.
(3.) J. Schanda, G. Schanda, and K. Muray, Light emitting diode standards, Proc. CIE 2nd Expert Symposium on LED Measurements, Gaithersburg, MD (2001) pp. 24-27.
(4.) C. C. Miller and Y. Ohno, Luminous intensity measurements of Light Emitting Diodes at NIST, Proc. CIE 2nd Expert Symposium on LED Measurements, Gaithersburg, MD (2001) pp. 28-32.
(5.) C. C. Miller and Y. Ohno, Total luminous flux calibrations of LEDs at NIST, Proc. Compound Semiconductor Manufacturing Exposition, Boston, MA, July 2001.
(6.) G. P. Eppeldauer, Spectral responsivity based calibration method of tristimulus colorimeters, J. Res. Natl. Inst. Stand. Technol. 103, 615-619 (1998).
(7.) S. W. Brown and B. C. Johnson, A portable integrating sphere source for the Earth Observing System's calibration validation program, accepted for publication in the Intl. J. Remote Sensing.
About the authors: Steven Brown is a physicist in the Optical Technology Division of the NIST Physics Laboratory. George Eppeldauer is an Electronics Engineer in the Optical Technology Division. Carlos Santana was in the NIST Summer Undergraduate Research Fellowship Program and is now a student at the University of Puerto Rico Founded in 1903, the University of Puerto Rico (Universidad de Puerto Rico in Spanish, UPR) is the oldest and largest university system in Puerto Rico. Though Puerto Rico is not a U.S. . The National Institute of Standards and Technology National Institute of Standards and Technology, governmental agency within the U.S. Dept. of Commerce with the mission of "working with industry to develop and apply technology, measurements, and standards" in the national interest. is an agency of the Technology Administration, U.S. Department of Commerce.